-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul 23 00:45:23 2021 UTC
-- File 20210627_232434_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.577695 to 38.927992
-each separately from 14s to 38s and from 47697us to 927992us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.469713 in 1 packets: 18
      10.879717 in 1 packets: 36
MIU packet time from 2021-06-27T23:24:48.577.695.000 to 2021-06-27T23:25:12.927.992.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.469713 in 1 packets: 18
      10.879717 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.9309404 to 39.9308530
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999491 in 1 packets: 18
      9.9999635 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      07 -> 45.9 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210627_232434_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5086354 to 16.5089244
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4696630  -- OK for SL --
   MIU time from 2021-06-27T23:24:48.577.695.000 to 2021-06-27T23:24:48.577.984.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.469713

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.9785874 to 29.9788764
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8796790  -- OK for SL --
   MIU time from 2021-06-27T23:25:02.047.697.000 to 2021-06-27T23:25:02.047.986.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.879717

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.8585554 to 40.8588444
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-27T23:25:12.927.703.000 to 2021-06-27T23:25:12.927.992.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.508635400 corrected by 40bit 10MHz counter of 15.930940400 obtained at 1Hz counter of 14. Result: 14.577697080
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-27T23:24:48.577.554.725
   Max 1st sample MIU time among all SL packets: 2021-06-27T23:25:12.927.851.725
   Min 1st sample XO time among all SL packets: 14.577556804
   Max 1st sample XO time among all SL packets: 38.927854558
   TM values from -286 to 273 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-27T23:24:48.577.554.725 (XO: 14.577556804 - 14.577862805, t=35.6 degC)  Rank: 00077
   TM values: Max =   261  Min =  -267  Ave =    -4.39  Std =    25.59
   Min TM value:-267 at an offset of 112.713 usec -> 2021-06-27T23:24:48.577.667.437 (XO: 14.577669517)
   Max TM value:261 at an offset of 115.338 usec -> 2021-06-27T23:24:48.577.670.062 (XO: 14.577672142)
   Max dTM/dt [1/us]:2.60e+03 at an offset of 115.213 usec -> 2021-06-27T23:24:48.577.669.937 (XO: 14.577672017)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 85.338 usec -> 2021-06-27T23:24:48.577.640.062 (XO: 14.577642142)

 - Block number: 2 starting at 2021-06-27T23:25:02.047.556.725 (XO: 28.047557796 - 28.047863797, t=35.6 degC)  Rank: 00078
   TM values: Max =   231  Min =  -286  Ave =    -4.45  Std =    25.20
   Min TM value:-286 at an offset of 117.350 usec -> 2021-06-27T23:25:02.047.674.075 (XO: 28.047675147)
   Max TM value:231 at an offset of 105.325 usec -> 2021-06-27T23:25:02.047.662.050 (XO: 28.047663122)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 114.925 usec -> 2021-06-27T23:25:02.047.671.650 (XO: 28.047672722)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 105.450 usec -> 2021-06-27T23:25:02.047.662.175 (XO: 28.047663247)

 - Block number: 3 starting at 2021-06-27T23:25:12.927.562.725 (XO: 38.927565557 - 38.927871558, t=35.6 degC)  Rank: 00086
   TM values: Max =   273  Min =  -272  Ave =    -4.45  Std =    25.60
   Min TM value:-272 at an offset of 111.225 usec -> 2021-06-27T23:25:12.927.673.950 (XO: 38.927676782)
   Max TM value:273 at an offset of 113.938 usec -> 2021-06-27T23:25:12.927.676.662 (XO: 38.927679495)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 113.813 usec -> 2021-06-27T23:25:12.927.676.537 (XO: 38.927679370)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 111.100 usec -> 2021-06-27T23:25:12.927.673.825 (XO: 38.927676657)
