-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jul 20 01:44:41 2021 UTC
-- File 20210627_175010_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.719790 to 24.001906
-each separately from 1s to 24s and from 1617us to 720079us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.281538 in 1 packets: 18
MIU packet time from 2021-06-27T17:50:11.719.790.000 to 2021-06-27T17:50:34.001.906.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.281538 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9134152 to 25.9133309
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999157 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210627_175010_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.6332050 to 3.6334940
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.2814550  -- OK for SL --
   MIU time from 2021-06-27T17:50:11.719.790.000 to 2021-06-27T17:50:11.720.079.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.281538

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.9149490 to 25.9152380
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-27T17:50:34.001.617.000 to 2021-06-27T17:50:34.001.906.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.633205000 corrected by 40bit 10MHz counter of 2.913415200 obtained at 1Hz counter of 1. Result: 1.719792463
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-27T17:50:11.719.649.725
   Max 1st sample MIU time among all SL packets: 2021-06-27T17:50:34.001.765.725
   Min 1st sample XO time among all SL packets: 1.719652188
   Max 1st sample XO time among all SL packets: 24.001766831
   TM values from -286 to 260 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-27T17:50:11.719.649.725 (XO: 1.719652188 - 1.719958189, t=35.6 degC)  Rank: 00075
   TM values: Max =   244  Min =  -286  Ave =    -4.46  Std =    23.60
   Min TM value:-286 at an offset of 107.875 usec -> 2021-06-27T17:50:11.719.757.600 (XO: 1.719760063)
   Max TM value:244 at an offset of 107.588 usec -> 2021-06-27T17:50:11.719.757.312 (XO: 1.719759776)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 110.288 usec -> 2021-06-27T17:50:11.719.760.012 (XO: 1.719762476)
   Min dTM/dt [1/us]:-2.55e+03 at an offset of 107.713 usec -> 2021-06-27T17:50:11.719.757.437 (XO: 1.719759901)

 - Block number: 2 starting at 2021-06-27T17:50:34.001.476.725 (XO: 24.001477830 - 24.001783831, t=35.6 degC)  Rank: 00085
   TM values: Max =   260  Min =  -236  Ave =    -4.47  Std =    27.35
   Min TM value:-236 at an offset of 122.238 usec -> 2021-06-27T17:50:34.001.598.962 (XO: 24.001600068)
   Max TM value:260 at an offset of 122.500 usec -> 2021-06-27T17:50:34.001.599.225 (XO: 24.001600331)
   Max dTM/dt [1/us]:2.32e+03 at an offset of 122.363 usec -> 2021-06-27T17:50:34.001.599.087 (XO: 24.001600193)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 107.563 usec -> 2021-06-27T17:50:34.001.584.287 (XO: 24.001585393)
