-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jul 20 00:43:48 2021 UTC
-- File 20210627_165506_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.638491 to 34.830122
-each separately from 4s to 34s and from 259045us to 830122us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.620265 in 1 packets: 18
      14.570499 in 1 packets: 36
MIU packet time from 2021-06-27T16:55:10.638.491.000 to 2021-06-27T16:55:40.830.122.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.620265 in 1 packets: 18
      14.570499 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9234309 to 35.9233206
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999412 in 1 packets: 18
      13.9999485 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210627_165506_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.5619216 to 6.5622106
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6202070  -- OK for SL --
   MIU time from 2021-06-27T16:55:10.638.491.000 to 2021-06-27T16:55:10.638.780.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.620265

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.1824176 to 22.1827066
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5704470  -- OK for SL --
   MIU time from 2021-06-27T16:55:26.259.045.000 to 2021-06-27T16:55:26.259.334.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.570499

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.7531536 to 36.7534426
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-27T16:55:40.829.833.000 to 2021-06-27T16:55:40.830.122.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.561921600 corrected by 40bit 10MHz counter of 5.923430900 obtained at 1Hz counter of 4. Result: 4.638493062
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-27T16:55:10.638.350.725
   Max 1st sample MIU time among all SL packets: 2021-06-27T16:55:40.829.981.725
   Min 1st sample XO time among all SL packets: 4.638352787
   Max 1st sample XO time among all SL packets: 34.829984713
   TM values from -296 to 269 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-27T16:55:10.638.350.725 (XO: 4.638352787 - 4.638658788, t=35.6 degC)  Rank: 00084
   TM values: Max =   269  Min =  -280  Ave =    -4.47  Std =    24.68
   Min TM value:-280 at an offset of 108.588 usec -> 2021-06-27T16:55:10.638.459.312 (XO: 4.638461375)
   Max TM value:269 at an offset of 124.225 usec -> 2021-06-27T16:55:10.638.474.950 (XO: 4.638477012)
   Max dTM/dt [1/us]:2.52e+03 at an offset of 124.113 usec -> 2021-06-27T16:55:10.638.474.837 (XO: 4.638476900)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 108.425 usec -> 2021-06-27T16:55:10.638.459.150 (XO: 4.638461212)

 - Block number: 2 starting at 2021-06-27T16:55:26.258.904.725 (XO: 20.258906183 - 20.259212184, t=35.6 degC)  Rank: 00081
   TM values: Max =   238  Min =  -273  Ave =    -4.44  Std =    31.13
   Min TM value:-273 at an offset of 166.176 usec -> 2021-06-27T16:55:26.259.070.900 (XO: 20.259072359)
   Max TM value:238 at an offset of 111.900 usec -> 2021-06-27T16:55:26.259.016.625 (XO: 20.259018083)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 111.788 usec -> 2021-06-27T16:55:26.259.016.512 (XO: 20.259017971)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 166.076 usec -> 2021-06-27T16:55:26.259.070.800 (XO: 20.259072259)

 - Block number: 3 starting at 2021-06-27T16:55:40.829.692.725 (XO: 34.829695712 - 34.830001713, t=35.6 degC)  Rank: 00078
   TM values: Max =   216  Min =  -296  Ave =    -4.44  Std =    19.68
   Min TM value:-296 at an offset of 112.775 usec -> 2021-06-27T16:55:40.829.805.500 (XO: 34.829808487)
   Max TM value:216 at an offset of 112.475 usec -> 2021-06-27T16:55:40.829.805.200 (XO: 34.829808187)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 112.875 usec -> 2021-06-27T16:55:40.829.805.600 (XO: 34.829808587)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 112.600 usec -> 2021-06-27T16:55:40.829.805.325 (XO: 34.829808312)
