-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jul 19 20:41:57 2021 UTC
-- File 20210627_124702_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.252650 to 5.252939
-each separately from 5s to 5s and from 252650us to 252939us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-06-27T12:47:07.252.650.000 to 2021-06-27T12:47:07.252.939.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9387674 to 6.9387674
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets (all packets)
Temperature XO: 
      BF -> 34.0 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210627_124702_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.1914183 to 7.1917073
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-27T12:47:07.252.650.000 to 2021-06-27T12:47:07.252.939.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.191418300 corrected by 40bit 10MHz counter of 6.938767400 obtained at 1Hz counter of 5. Result: 5.252651784
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-06-27T12:47:07.252.509.725
   Max 1st sample MIU time among all SL packets: 2021-06-27T12:47:07.252.798.725
   Min 1st sample XO time among all SL packets: 5.252511509
   Max 1st sample XO time among all SL packets: 5.252800510
   TM values from -270 to 246 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-27T12:47:07.252.509.725 (XO: 5.252511509 - 5.252817510, t=34.0 degC)  Rank: 00080
   TM values: Max =   246  Min =  -270  Ave =    -4.44  Std =    27.34
   Min TM value:-270 at an offset of 110.050 usec -> 2021-06-27T12:47:07.252.619.775 (XO: 5.252621559)
   Max TM value:246 at an offset of 125.363 usec -> 2021-06-27T12:47:07.252.635.087 (XO: 5.252636872)
   Max dTM/dt [1/us]:2.46e+03 at an offset of 125.238 usec -> 2021-06-27T12:47:07.252.634.962 (XO: 5.252636747)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 109.900 usec -> 2021-06-27T12:47:07.252.619.625 (XO: 5.252621409)
