-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jul 19 03:47:18 2021 UTC
-- File 20210618_201602_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.094383 to 25.268051
-each separately from 2s to 25s and from 94383us to 268051us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.173090 in 1 packets: 18
MIU packet time from 2021-06-18T20:16:06.094.383.000 to 2021-06-18T20:16:29.268.051.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.173090 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8816660 to 26.8815835
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999175 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210618_201602_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.9760505 to 3.9763395
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.1730070  -- OK for SL --
   MIU time from 2021-06-18T20:16:06.094.383.000 to 2021-06-18T20:16:06.094.672.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.173090

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.1493465 to 27.1496355
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-18T20:16:29.267.762.000 to 2021-06-18T20:16:29.268.051.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.976050500 corrected by 40bit 10MHz counter of 3.881666000 obtained at 1Hz counter of 2. Result: 2.094384840
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-18T20:16:06.094.242.725
   Max 1st sample MIU time among all SL packets: 2021-06-18T20:16:29.267.910.725
   Min 1st sample XO time among all SL packets: 2.094244564
   Max 1st sample XO time among all SL packets: 25.267912689
   TM values from -162 to 155 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-18T20:16:06.094.242.725 (XO: 2.094244564 - 2.094550565, t=35.6 degC)  Rank: 00057
   TM values: Max =   120  Min =  -139  Ave =    -4.41  Std =    37.35
   Min TM value:-139 at an offset of 274.776 usec -> 2021-06-18T20:16:06.094.517.500 (XO: 2.094519340)
   Max TM value:120 at an offset of 122.375 usec -> 2021-06-18T20:16:06.094.365.100 (XO: 2.094366940)
   Max dTM/dt [1/us]:1.37e+03 at an offset of 177.463 usec -> 2021-06-18T20:16:06.094.420.188 (XO: 2.094422027)
   Min dTM/dt [1/us]:-1.32e+03 at an offset of 177.101 usec -> 2021-06-18T20:16:06.094.419.825 (XO: 2.094421665)

 - Block number: 2 starting at 2021-06-18T20:16:29.267.621.725 (XO: 25.267623688 - 25.267929690, t=35.6 degC)  Rank: 00068
   TM values: Max =   155  Min =  -162  Ave =    -4.44  Std =    58.15
   Min TM value:-162 at an offset of 113.388 usec -> 2021-06-18T20:16:29.267.735.112 (XO: 25.267737076)
   Max TM value:155 at an offset of 211.288 usec -> 2021-06-18T20:16:29.267.833.013 (XO: 25.267834977)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 120.475 usec -> 2021-06-18T20:16:29.267.742.200 (XO: 25.267744164)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 113.975 usec -> 2021-06-18T20:16:29.267.735.700 (XO: 25.267737664)
