-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jul 19 00:55:52 2021 UTC
-- File 20210617_053527_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.311427 to 61.944068
-each separately from 3s to 61s and from 238669us to 978766us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.666761 in 1 packets: 18
      17.259903 in 1 packets: 36
      10.496627 in 1 packets: 54
      14.207905 in 1 packets: 72
MIU packet time from 2021-06-17T05:35:32.311.427.000 to 2021-06-17T05:36:30.944.068.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.666761 in 1 packets: 18
      17.259903 in 1 packets: 36
      10.496627 in 1 packets: 54
      14.207905 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4505260 to 62.4503189
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999429 in 1 packets: 18
      17.9999357 in 1 packets: 36
      9.9999643 in 1 packets: 54
      13.9999500 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 90 packets (all packets)
Number of generated alerts: 
      0 in 90 packets (all packets)
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210617_053527_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      45 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.7619541 to 4.7622431
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6667030  -- OK for SL --
   MIU time from 2021-06-17T05:35:32.311.427.000 to 2021-06-17T05:35:32.311.716.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.666761

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      46 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.4289461 to 21.4292351
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.2598390  -- OK for SL --
   MIU time from 2021-06-17T05:35:48.978.477.000 to 2021-06-17T05:35:48.978.766.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.259903

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      47 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.6890741 to 38.6893631
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.4965910  -- OK for SL --
   MIU time from 2021-06-17T05:36:06.238.669.000 to 2021-06-17T05:36:06.238.958.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.496627

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.1859541 to 49.1862431
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2078550  -- OK for SL --
   MIU time from 2021-06-17T05:36:16.735.585.000 to 2021-06-17T05:36:16.735.874.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.207905

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.3940981 to 63.3943871
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-17T05:36:30.943.779.000 to 2021-06-17T05:36:30.944.068.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.761954100 corrected by 40bit 10MHz counter of 4.450526000 obtained at 1Hz counter of 3. Result: 3.311429221
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-06-17T05:35:32.311.286.725
   Max 1st sample MIU time among all SL packets: 2021-06-17T05:36:30.943.927.725
   Min 1st sample XO time among all SL packets: 3.311288946
   Max 1st sample XO time among all SL packets: 61.943931323
   TM values from -203 to 193 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-17T05:35:32.311.286.725 (XO: 3.311288946 - 3.311594947, t=35.6 degC)  Rank: 00045
   TM values: Max =   135  Min =  -149  Ave =    -4.51  Std =    48.42
   Min TM value:-149 at an offset of 131.125 usec -> 2021-06-17T05:35:32.311.417.850 (XO: 3.311420071)
   Max TM value:135 at an offset of 131.088 usec -> 2021-06-17T05:35:32.311.417.812 (XO: 3.311420034)
   Max dTM/dt [1/us]:9.95e+02 at an offset of 173.538 usec -> 2021-06-17T05:35:32.311.460.263 (XO: 3.311462484)
   Min dTM/dt [1/us]:-9.54e+02 at an offset of 194.026 usec -> 2021-06-17T05:35:32.311.480.750 (XO: 3.311482971)

 - Block number: 2 starting at 2021-06-17T05:35:48.978.336.725 (XO: 19.978340349 - 19.978646350, t=35.6 degC)  Rank: 00046
   TM values: Max =   183  Min =  -185  Ave =    -4.44  Std =    59.31
   Min TM value:-185 at an offset of 112.888 usec -> 2021-06-17T05:35:48.978.449.612 (XO: 19.978453237)
   Max TM value:183 at an offset of 113.438 usec -> 2021-06-17T05:35:48.978.450.162 (XO: 19.978453787)
   Max dTM/dt [1/us]:1.36e+03 at an offset of 141.838 usec -> 2021-06-17T05:35:48.978.478.562 (XO: 19.978482187)
   Min dTM/dt [1/us]:-1.36e+03 at an offset of 130.663 usec -> 2021-06-17T05:35:48.978.467.387 (XO: 19.978471012)

 - Block number: 3 starting at 2021-06-17T05:36:06.238.528.725 (XO: 37.238530060 - 37.238836061, t=35.6 degC)  Rank: 00047
   TM values: Max =   137  Min =  -155  Ave =    -4.52  Std =    49.61
   Min TM value:-155 at an offset of 148.501 usec -> 2021-06-17T05:36:06.238.677.225 (XO: 37.238678560)
   Max TM value:137 at an offset of 149.051 usec -> 2021-06-17T05:36:06.238.677.775 (XO: 37.238679110)
   Max dTM/dt [1/us]:9.48e+02 at an offset of 112.313 usec -> 2021-06-17T05:36:06.238.641.037 (XO: 37.238642373)
   Min dTM/dt [1/us]:-1.03e+03 at an offset of 305.989 usec -> 2021-06-17T05:36:06.238.834.713 (XO: 37.238836048)

 - Block number: 4 starting at 2021-06-17T05:36:16.735.444.725 (XO: 47.735447573 - 47.735753574, t=35.6 degC)  Rank: 00078
   TM values: Max =   174  Min =  -185  Ave =    -4.49  Std =    69.97
   Min TM value:-185 at an offset of 130.338 usec -> 2021-06-17T05:36:16.735.575.062 (XO: 47.735577911)
   Max TM value:174 at an offset of 129.575 usec -> 2021-06-17T05:36:16.735.574.300 (XO: 47.735577148)
   Max dTM/dt [1/us]:1.10e+03 at an offset of 172.751 usec -> 2021-06-17T05:36:16.735.617.475 (XO: 47.735620323)
   Min dTM/dt [1/us]:-1.09e+03 at an offset of 149.638 usec -> 2021-06-17T05:36:16.735.594.363 (XO: 47.735597211)

 - Block number: 5 starting at 2021-06-17T05:36:30.943.638.725 (XO: 61.943642322 - 61.943948323, t=35.6 degC)  Rank: 00063
   TM values: Max =   193  Min =  -203  Ave =    -4.55  Std =    74.94
   Min TM value:-203 at an offset of 105.738 usec -> 2021-06-17T05:36:30.943.744.462 (XO: 61.943748060)
   Max TM value:193 at an offset of 129.575 usec -> 2021-06-17T05:36:30.943.768.300 (XO: 61.943771898)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 305.989 usec -> 2021-06-17T05:36:30.943.944.713 (XO: 61.943948311)
   Min dTM/dt [1/us]:-1.27e+03 at an offset of 114.925 usec -> 2021-06-17T05:36:30.943.753.650 (XO: 61.943757248)
