-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul 18 20:52:14 2021 UTC
-- File 20210614_200525_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177028
-each separately from 0s to 14s and from 0us to 841255us.
   Differences in neigboring packets: 
      0.833940 in 2 packets: 1, 7
      0.833943 in 9 packets: 2, 4, 5, 9, 10, 12, 14, 16, 17
      0.833944 in 5 packets: 3, 6, 8, 11, 15
      0.833941 in 1 packets: 13
MIU packet time from 2021-06-14T20:05:27.000.000.000 to 2021-06-14T20:05:41.177.028.000
   Differences in neigboring packets: 
      0.833940 in 2 packets: 1, 7
      0.833943 in 9 packets: 2, 4, 5, 9, 10, 12, 14, 16, 17
      0.833944 in 5 packets: 3, 6, 8, 11, 15
      0.833941 in 1 packets: 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7914023 to 15.7913530
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999965 in 11 packets: 2, 3, 4, 6, 8, 9, 10, 12, 14, 15, 17
      0.9999964 in 3 packets: 5, 11, 16
   XO PPM: 
      -3.5 in 11 packets: 2, 3, 4, 6, 8, 9, 10, 12, 14, 15, 17
      -3.6 in 3 packets: 5, 11, 16
Temperature FPGA: 
      04 -> 49.8 degC in 3 packets: 1, 11, 12
      05 -> 48.5 degC in 13 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 15, 16, 17, 18
      06 -> 47.2 degC in 2 packets: 13, 14
Temperature XO: 
      C0 -> 35.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210614_200525_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7914024 to 15.9683824
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-14T20:05:27.000.000.000 to 2021-06-14T20:05:41.177.028.000
      Difference between neigboring packets: 
      0.833940 in 2 packets: 1, 7
      0.833943 in 9 packets: 2, 4, 5, 9, 10, 12, 14, 16, 17
      0.833944 in 5 packets: 3, 6, 8, 11, 15
      0.833941 in 1 packets: 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.791402400 corrected by 40bit 10MHz counter of 1.791402300 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SR packets: 2021-06-14T20:05:26.999.999.925
   Max 1st sample MIU time among all SR packets: 2021-06-14T20:05:41.177.027.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177029945
   TM values from -167 to 163 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2021-06-14T20:05:26.999.999.925 (XO: 0.000000025 - 0.000017025, t=35.6 degC)
   TM values: Max =    53  Min =   -62  Ave =    -4.17  Std =    22.59

 - Packet number: 2 starting at 2021-06-14T20:05:27.833.939.925 (XO: 0.833942944 - 0.833959944, t=35.6 degC)
   TM values: Max =    76  Min =   -95  Ave =    -4.75  Std =    34.94

 - Packet number: 3 starting at 2021-06-14T20:05:28.667.882.925 (XO: 1.667885863 - 1.667902863, t=35.6 degC)
   TM values: Max =   113  Min =  -113  Ave =    -4.15  Std =    46.60

 - Packet number: 4 starting at 2021-06-14T20:05:29.501.826.925 (XO: 2.501828781 - 2.501845781, t=35.6 degC)
   TM values: Max =   124  Min =  -125  Ave =    -4.59  Std =    48.58

 - Packet number: 5 starting at 2021-06-14T20:05:30.335.769.925 (XO: 3.335771734 - 3.335788734, t=35.6 degC)
   TM values: Max =   109  Min =  -115  Ave =    -4.38  Std =    44.56

 - Packet number: 6 starting at 2021-06-14T20:05:31.169.712.925 (XO: 4.169714719 - 4.169731719, t=35.6 degC)
   TM values: Max =    90  Min =   -89  Ave =    -4.28  Std =    31.52

 - Packet number: 7 starting at 2021-06-14T20:05:32.003.656.925 (XO: 5.003657638 - 5.003674638, t=35.6 degC)
   TM values: Max =    76  Min =   -86  Ave =    -4.54  Std =    27.61

 - Packet number: 8 starting at 2021-06-14T20:05:32.837.596.925 (XO: 5.837600557 - 5.837617557, t=35.6 degC)
   TM values: Max =   125  Min =  -130  Ave =    -4.58  Std =    47.20

 - Packet number: 9 starting at 2021-06-14T20:05:33.671.540.925 (XO: 6.671543475 - 6.671560475, t=35.6 degC)
   TM values: Max =   106  Min =  -106  Ave =    -4.32  Std =    46.28

 - Packet number: 10 starting at 2021-06-14T20:05:34.505.483.925 (XO: 7.505486394 - 7.505503394, t=35.6 degC)
   TM values: Max =    98  Min =   -93  Ave =    -4.61  Std =    34.14

 - Packet number: 11 starting at 2021-06-14T20:05:35.339.426.925 (XO: 8.339429347 - 8.339446347, t=35.6 degC)
   TM values: Max =   142  Min =  -146  Ave =    -4.44  Std =    57.38

 - Packet number: 12 starting at 2021-06-14T20:05:36.173.370.925 (XO: 9.173372332 - 9.173389332, t=35.6 degC)
   TM values: Max =    96  Min =   -94  Ave =    -4.42  Std =    31.60

 - Packet number: 13 starting at 2021-06-14T20:05:37.007.313.925 (XO: 10.007315251 - 10.007332251, t=35.6 degC)
   TM values: Max =    86  Min =  -106  Ave =    -4.42  Std =    33.15

 - Packet number: 14 starting at 2021-06-14T20:05:37.841.254.925 (XO: 10.841258169 - 10.841275169, t=35.6 degC)
   TM values: Max =   145  Min =  -158  Ave =    -4.56  Std =    63.90

 - Packet number: 15 starting at 2021-06-14T20:05:38.675.197.925 (XO: 11.675201088 - 11.675218088, t=35.6 degC)
   TM values: Max =   106  Min =  -122  Ave =    -4.28  Std =    44.01

 - Packet number: 16 starting at 2021-06-14T20:05:39.509.141.925 (XO: 12.509144058 - 12.509161058, t=35.6 degC)
   TM values: Max =   163  Min =  -167  Ave =    -4.45  Std =    64.29

 - Packet number: 17 starting at 2021-06-14T20:05:40.343.084.925 (XO: 13.343087026 - 13.343104026, t=35.6 degC)
   TM values: Max =   122  Min =  -129  Ave =    -4.40  Std =    46.37

 - Packet number: 18 starting at 2021-06-14T20:05:41.177.027.925 (XO: 14.177029945 - 14.177046945, t=35.6 degC)
   TM values: Max =    76  Min =   -87  Ave =    -4.37  Std =    28.02
