-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jun 10 23:07:35 2021 UTC
-- File 20210610_123014_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.960226 to 29.629640
-each separately from 5s to 29s and from 629351us to 960515us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.668836 in 1 packets: 18
MIU packet time from 2021-06-10T12:30:22.960.226.000 to 2021-06-10T12:30:46.629.640.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.668836 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8547720 to 30.8546870
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999150 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210610_123014_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      45 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8149982 to 7.8152872
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.6687510  -- OK for SL --
   MIU time from 2021-06-10T12:30:22.960.226.000 to 2021-06-10T12:30:22.960.515.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.668836

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      100 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4840382 to 31.4843272
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-10T12:30:46.629.351.000 to 2021-06-10T12:30:46.629.640.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.814998200 corrected by 40bit 10MHz counter of 6.854772000 obtained at 1Hz counter of 5. Result: 5.960229657
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-06-10T12:30:22.960.085.725
   Max 1st sample MIU time among all SL packets: 2021-06-10T12:30:46.629.499.725
   Min 1st sample XO time among all SL packets: 5.960089381
   Max 1st sample XO time among all SL packets: 29.629502128
   TM values from -453 to 230 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-10T12:30:22.960.085.725 (XO: 5.960089381 - 5.960395382, t=35.6 degC)  Rank: 00045
   TM values: Max =   204  Min =  -453  Ave =    -4.61  Std =    45.02
   Min TM value:-453 at an offset of 236.126 usec -> 2021-06-10T12:30:22.960.321.850 (XO: 5.960325507)
   Max TM value:204 at an offset of 287.026 usec -> 2021-06-10T12:30:22.960.372.751 (XO: 5.960376407)
   Max dTM/dt [1/us]:2.94e+03 at an offset of 275.588 usec -> 2021-06-10T12:30:22.960.361.313 (XO: 5.960364970)
   Min dTM/dt [1/us]:-3.30e+03 at an offset of 275.463 usec -> 2021-06-10T12:30:22.960.361.188 (XO: 5.960364845)

 - Block number: 2 starting at 2021-06-10T12:30:46.629.210.725 (XO: 29.629213127 - 29.629519128, t=35.6 degC)  Rank: 00100
   TM values: Max =   230  Min =  -447  Ave =    -4.51  Std =    27.88
   Min TM value:-447 at an offset of 106.988 usec -> 2021-06-10T12:30:46.629.317.712 (XO: 29.629320115)
   Max TM value:230 at an offset of 107.550 usec -> 2021-06-10T12:30:46.629.318.275 (XO: 29.629320678)
   Max dTM/dt [1/us]:2.77e+03 at an offset of 107.063 usec -> 2021-06-10T12:30:46.629.317.787 (XO: 29.629320190)
   Min dTM/dt [1/us]:-3.36e+03 at an offset of 107.825 usec -> 2021-06-10T12:30:46.629.318.550 (XO: 29.629320953)
