-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jun  7 17:42:51 2021 UTC
-- File 20210604_183139_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.337362 to 66.849497
-each separately from 2s to 66s and from 337362us to 903626us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.157632 in 1 packets: 18
      15.407765 in 1 packets: 36
      0.849171 in 1 packets: 54
      21.096122 in 1 packets: 72
MIU packet time from 2021-06-04T18:31:44.337.362.000 to 2021-06-04T18:32:48.849.497.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.157632 in 1 packets: 18
      15.407765 in 1 packets: 36
      0.849171 in 1 packets: 54
      21.096122 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4886179 to 67.4883881
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.9999032 in 1 packets: 18
      14.9999461 in 1 packets: 36
      0.9999964 in 1 packets: 54
      20.9999245 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C0 -> 35.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210604_183139_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      49 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.8259804 to 3.8262694
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.1575350  -- OK for SL --
   MIU time from 2021-06-04T18:31:44.337.362.000 to 2021-06-04T18:31:44.337.651.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.157632

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      51 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.9838044 to 30.9840934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4077110  -- OK for SL --
   MIU time from 2021-06-04T18:32:11.495.283.000 to 2021-06-04T18:32:11.495.572.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.407765

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.3918044 to 46.3920934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.8491670  -- OK for SL --
   MIU time from 2021-06-04T18:32:26.903.337.000 to 2021-06-04T18:32:26.903.626.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.849171

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.2412604 to 47.2415494
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0960470  -- OK for SL --
   MIU time from 2021-06-04T18:32:27.752.797.000 to 2021-06-04T18:32:27.753.086.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.096122

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.3375964 to 68.3378854
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-04T18:32:48.849.208.000 to 2021-06-04T18:32:48.849.497.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.825980400 corrected by 40bit 10MHz counter of 3.488617900 obtained at 1Hz counter of 2. Result: 2.337363715
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-04T18:31:44.337.221.725
   Max 1st sample MIU time among all SL packets: 2021-06-04T18:32:48.849.356.725
   Min 1st sample XO time among all SL packets: 2.337223439
   Max 1st sample XO time among all SL packets: 66.849360083
   TM values from -283 to 283 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-04T18:31:44.337.221.725 (XO: 2.337223439 - 2.337529440, t=35.6 degC)  Rank: 00049
   TM values: Max =   213  Min =  -218  Ave =    -4.37  Std =    60.83
   Min TM value:-218 at an offset of 169.151 usec -> 2021-06-04T18:31:44.337.390.875 (XO: 2.337392590)
   Max TM value:213 at an offset of 123.975 usec -> 2021-06-04T18:31:44.337.345.700 (XO: 2.337347414)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 142.126 usec -> 2021-06-04T18:31:44.337.363.850 (XO: 2.337365565)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 137.688 usec -> 2021-06-04T18:31:44.337.359.412 (XO: 2.337361127)

 - Block number: 2 starting at 2021-06-04T18:32:11.495.142.725 (XO: 29.495144808 - 29.495450809, t=35.6 degC)  Rank: 00051
   TM values: Max =   163  Min =  -188  Ave =    -4.31  Std =    55.28
   Min TM value:-188 at an offset of 133.563 usec -> 2021-06-04T18:32:11.495.276.287 (XO: 29.495278371)
   Max TM value:163 at an offset of 134.038 usec -> 2021-06-04T18:32:11.495.276.762 (XO: 29.495278846)
   Max dTM/dt [1/us]:8.95e+02 at an offset of 144.788 usec -> 2021-06-04T18:32:11.495.287.513 (XO: 29.495289596)
   Min dTM/dt [1/us]:-8.43e+02 at an offset of 122.788 usec -> 2021-06-04T18:32:11.495.265.512 (XO: 29.495267595)

 - Block number: 3 starting at 2021-06-04T18:32:26.903.196.725 (XO: 44.903200177 - 44.903506178, t=35.6 degC)  Rank: 00076
   TM values: Max =   215  Min =  -233  Ave =    -4.41  Std =    73.90
   Min TM value:-233 at an offset of 121.313 usec -> 2021-06-04T18:32:26.903.318.037 (XO: 44.903321489)
   Max TM value:215 at an offset of 127.988 usec -> 2021-06-04T18:32:26.903.324.712 (XO: 44.903328164)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 128.900 usec -> 2021-06-04T18:32:26.903.325.625 (XO: 44.903329077)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 122.225 usec -> 2021-06-04T18:32:26.903.318.950 (XO: 44.903322402)

 - Block number: 4 starting at 2021-06-04T18:32:27.752.656.725 (XO: 45.752659235 - 45.752965236, t=35.6 degC)  Rank: 00073
   TM values: Max =   217  Min =  -249  Ave =    -4.35  Std =    75.62
   Min TM value:-249 at an offset of 123.500 usec -> 2021-06-04T18:32:27.752.780.225 (XO: 45.752782735)
   Max TM value:217 at an offset of 129.850 usec -> 2021-06-04T18:32:27.752.786.575 (XO: 45.752789085)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 137.150 usec -> 2021-06-04T18:32:27.752.793.875 (XO: 45.752796385)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 123.963 usec -> 2021-06-04T18:32:27.752.780.687 (XO: 45.752783198)

 - Block number: 5 starting at 2021-06-04T18:32:48.849.067.725 (XO: 66.849071082 - 66.849377083, t=35.6 degC)  Rank: 00082
   TM values: Max =   283  Min =  -283  Ave =    -4.40  Std =    85.68
   Min TM value:-283 at an offset of 131.113 usec -> 2021-06-04T18:32:48.849.198.837 (XO: 66.849202195)
   Max TM value:283 at an offset of 151.651 usec -> 2021-06-04T18:32:48.849.219.375 (XO: 66.849222732)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 305.989 usec -> 2021-06-04T18:32:48.849.373.713 (XO: 66.849377070)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 151.251 usec -> 2021-06-04T18:32:48.849.218.975 (XO: 66.849222332)
