-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 12 14:31:54 2021 UTC
-- File 20200923_042500_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.499437 to 15.307159
-each separately from 4s to 15s and from 306870us to 499726us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.807144 in 1 packets: 18
MIU packet time from 2020-09-23T04:25:07.499.437.000 to 2020-09-23T04:25:18.307.159.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.807144 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.7876325 to 16.7875911
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999586 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20200923_042500_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      59 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.2870697 to 6.2873587
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8071030  -- OK for SL --
   MIU time from 2020-09-23T04:25:07.499.437.000 to 2020-09-23T04:25:07.499.726.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.807144

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0944617 to 17.0947507
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2020-09-23T04:25:18.306.870.000 to 2020-09-23T04:25:18.307.159.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.287069700 corrected by 40bit 10MHz counter of 5.787632500 obtained at 1Hz counter of 4. Result: 4.499439048
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2020-09-23T04:25:07.499.296.725
   Max 1st sample MIU time among all SL packets: 2020-09-23T04:25:18.307.018.725
   Min 1st sample XO time among all SL packets: 4.499298772
   Max 1st sample XO time among all SL packets: 15.307020461
   TM values from -232 to 438 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2020-09-23T04:25:07.499.296.725 (XO: 4.499298772 - 4.499604774, t=35.6 degC)  Rank: 00059
   TM values: Max =   156  Min =  -123  Ave =    -4.42  Std =    18.09
   Min TM value:-123 at an offset of 142.026 usec -> 2020-09-23T04:25:07.499.438.750 (XO: 4.499440798)
   Max TM value:156 at an offset of 124.000 usec -> 2020-09-23T04:25:07.499.420.725 (XO: 4.499422773)
   Max dTM/dt [1/us]:1.04e+03 at an offset of 126.188 usec -> 2020-09-23T04:25:07.499.422.912 (XO: 4.499424960)
   Min dTM/dt [1/us]:-1.26e+03 at an offset of 124.125 usec -> 2020-09-23T04:25:07.499.420.850 (XO: 4.499422898)

 - Block number: 2 starting at 2020-09-23T04:25:18.306.729.725 (XO: 15.306731460 - 15.307037461, t=35.6 degC)  Rank: 00088
   TM values: Max =   438  Min =  -232  Ave =    -4.55  Std =    22.81
   Min TM value:-232 at an offset of 106.238 usec -> 2020-09-23T04:25:18.306.835.962 (XO: 15.306837698)
   Max TM value:438 at an offset of 107.188 usec -> 2020-09-23T04:25:18.306.836.912 (XO: 15.306838648)
   Max dTM/dt [1/us]:2.75e+03 at an offset of 107.075 usec -> 2020-09-23T04:25:18.306.836.800 (XO: 15.306838535)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 106.038 usec -> 2020-09-23T04:25:18.306.835.762 (XO: 15.306837498)
