-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 12 03:20:49 2021 UTC
-- File 20200922_115502_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.993072 to 16.163485
-each separately from 8s to 16s and from 163196us to 993361us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.169835 in 1 packets: 18
MIU packet time from 2020-09-22T11:55:13.993.072.000 to 2020-09-22T11:55:21.163.485.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.169835 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4698762 to 17.4698460
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999698 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20200922_115502_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.4629474 to 10.4632364
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.1698070  -- OK for SL --
   MIU time from 2020-09-22T11:55:13.993.072.000 to 2020-09-22T11:55:13.993.361.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.169835

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      139 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.6330434 to 17.6333324
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2020-09-22T11:55:21.163.196.000 to 2020-09-22T11:55:21.163.485.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.462947400 corrected by 40bit 10MHz counter of 9.469876200 obtained at 1Hz counter of 8. Result: 8.993074974
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2020-09-22T11:55:13.992.931.725
   Max 1st sample MIU time among all SL packets: 2020-09-22T11:55:21.163.344.725
   Min 1st sample XO time among all SL packets: 8.992934698
   Max 1st sample XO time among all SL packets: 16.163346729
   TM values from -141 to 193 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2020-09-22T11:55:13.992.931.725 (XO: 8.992934698 - 8.993240699, t=35.6 degC)  Rank: 00078
   TM values: Max =    95  Min =   -82  Ave =    -4.54  Std =    12.55
   Min TM value:-82 at an offset of 49.225 usec -> 2020-09-22T11:55:13.992.980.950 (XO: 8.992983923)
   Max TM value:95 at an offset of 43.925 usec -> 2020-09-22T11:55:13.992.975.650 (XO: 8.992978623)
   Max dTM/dt [1/us]:1.40e+02 at an offset of 110.338 usec -> 2020-09-22T11:55:13.993.042.062 (XO: 8.993045036)
   Min dTM/dt [1/us]:-1.23e+02 at an offset of 58.600 usec -> 2020-09-22T11:55:13.992.990.325 (XO: 8.992993298)

 - Block number: 2 starting at 2020-09-22T11:55:21.163.055.725 (XO: 16.163057728 - 16.163363729, t=35.6 degC)  Rank: 00139
   TM values: Max =   193  Min =  -141  Ave =    -4.45  Std =    22.62
   Min TM value:-141 at an offset of 119.675 usec -> 2020-09-22T11:55:21.163.175.400 (XO: 16.163177404)
   Max TM value:193 at an offset of 114.588 usec -> 2020-09-22T11:55:21.163.170.312 (XO: 16.163172316)
   Max dTM/dt [1/us]:4.05e+02 at an offset of 113.438 usec -> 2020-09-22T11:55:21.163.169.162 (XO: 16.163171166)
   Min dTM/dt [1/us]:-4.18e+02 at an offset of 140.613 usec -> 2020-09-22T11:55:21.163.196.338 (XO: 16.163198341)
