-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 12 02:05:23 2021 UTC
-- File 20200922_105438_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.656940 to 24.894237
-each separately from 3s to 24s and from 656940us to 894237us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.236719 in 1 packets: 18
MIU packet time from 2020-09-22T10:54:44.656.940.000 to 2020-09-22T10:55:05.894.237.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.236719 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7488448 to 25.7487658
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999210 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20200922_105438_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.4057852 to 5.4060742
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2366390  -- OK for SL --
   MIU time from 2020-09-22T10:54:44.656.940.000 to 2020-09-22T10:54:44.657.229.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.236719

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      110 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6427132 to 26.6430022
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2020-09-22T10:55:05.893.948.000 to 2020-09-22T10:55:05.894.237.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.405785200 corrected by 40bit 10MHz counter of 4.748844800 obtained at 1Hz counter of 3. Result: 3.656942831
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2020-09-22T10:54:44.656.799.725
   Max 1st sample MIU time among all SL packets: 2020-09-22T10:55:05.894.096.725
   Min 1st sample XO time among all SL packets: 3.656802555
   Max 1st sample XO time among all SL packets: 24.894099523
   TM values from -344 to 543 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2020-09-22T10:54:44.656.799.725 (XO: 3.656802555 - 3.657108556, t=35.6 degC)  Rank: 00071
   TM values: Max =   167  Min =  -326  Ave =    -4.45  Std =    19.53
   Min TM value:-326 at an offset of 123.188 usec -> 2020-09-22T10:54:44.656.922.912 (XO: 3.656925743)
   Max TM value:167 at an offset of 24.713 usec -> 2020-09-22T10:54:44.656.824.437 (XO: 3.656827268)
   Max dTM/dt [1/us]:3.93e+03 at an offset of 24.650 usec -> 2020-09-22T10:54:44.656.824.375 (XO: 3.656827205)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 121.300 usec -> 2020-09-22T10:54:44.656.921.025 (XO: 3.656923856)

 - Block number: 2 starting at 2020-09-22T10:55:05.893.807.725 (XO: 24.893810521 - 24.894116523, t=35.6 degC)  Rank: 00110
   TM values: Max =   543  Min =  -344  Ave =    -4.62  Std =    33.42
   Min TM value:-344 at an offset of 105.438 usec -> 2020-09-22T10:55:05.893.913.162 (XO: 24.893915959)
   Max TM value:543 at an offset of 104.863 usec -> 2020-09-22T10:55:05.893.912.587 (XO: 24.893915384)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 104.800 usec -> 2020-09-22T10:55:05.893.912.525 (XO: 24.893915322)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 104.938 usec -> 2020-09-22T10:55:05.893.912.662 (XO: 24.893915459)
