-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20200604_150115_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 17 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Hz and MHz counters from 4.665412 to 4.665701
-each separately from 4s to 4s and from 665412us to 665701us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2020-06-04T15:01:22.665.412.000 to 2020-06-04T15:01:22.665.701.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.7523279 to 5.7523279
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      04 -> 49.8°C in 18 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
Zero: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20200604_150115_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      54 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.4177401 to 6.4180291
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2020-06-04T15:01:22.665.412.000 to 2020-06-04T15:01:22.665.701.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2020-06-04T15:01:22.665.271.750
   Max SL corrected packet time: 2020-06-04T15:01:22.665.560.750
   Min SL corrected XO rel time: 4.6654145
   Max SL corrected XO rel time: 4.6657035
   TM values from -155 to 383 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2020-06-04T15:01:22.665.271.750 (XO: 4.6654145, t=35.6°C)  Rank: 00054
   TM values: Max =   383  Min =  -155  Ave =    -4.66  Std =    18.77
   Min TM value:-155 at an offset of 37.575 usec -> 2020-06-04T15:01:22.665.309.325 (XO: 4.6654520)
   Max TM value:383 at an offset of 49.163 usec -> 2020-06-04T15:01:22.665.320.912 (XO: 4.6654636)
   Max dTM/dt [1/us]::3.08e+03 at an offset of 49.125 usec -> 2020-06-04T15:01:22.665.320.875 (XO: 4.6654636)
   Min dTM/dt [1/us]::-2.81e+03 at an offset of 49.225 usec -> 2020-06-04T15:01:22.665.320.975 (XO: 4.6654637)
