-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180731_030053_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.349756 to 32.490790
-each separately from 1s to 32s and from 349756us to 982565us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.632231 in 1 packets - #/No: 18/18
      16.507936 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-31T03:00:54.349.756.000 to 2018-07-31T03:01:25.490.790.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.632231 in 1 packets - #/No: 18/18
      16.507936 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.3795774 to 33.3794569
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999456 in 1 packets - #/No: 18/18
      16.9999339 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180731_030053_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.7293334 to 2.7296224
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6321750  -- OK for SL --
   MIU time from 2018-07-31T03:00:54.349.756.000 to 2018-07-31T03:00:54.350.045.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.632231

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3617974 to 17.3620864
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5078710  -- OK for SL --
   MIU time from 2018-07-31T03:01:08.982.276.000 to 2018-07-31T03:01:08.982.565.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.507936

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.8699574 to 33.8702464
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-31T03:01:25.490.501.000 to 2018-07-31T03:01:25.490.790.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-31T03:00:54.349.615.750
   Max SL corrected packet time: 2018-07-31T03:01:25.490.649.750
   Min SL corrected XO rel time: 1.3497574
   Max SL corrected XO rel time: 32.4907914
   TM values from -16 to 7 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-31T03:00:54.349.615.750 (XO: 1.3497574, t=35.6°C)  Rank: 00004
   TM values: Max =     7  Min =   -16  Ave =    -4.37  Std =     2.85
   Min TM value:-16 at an offset of 114.950 usec -> 2018-07-31T03:00:54.349.730.700 (XO: 1.3498723)
   Max TM value:7 at an offset of 116.263 usec -> 2018-07-31T03:00:54.349.732.012 (XO: 1.3498736)
   Max dTM/dt [1/us]::9.30e+01 at an offset of 115.825 usec -> 2018-07-31T03:00:54.349.731.575 (XO: 1.3498732)
   Min dTM/dt [1/us]::-8.30e+01 at an offset of 10.238 usec -> 2018-07-31T03:00:54.349.625.987 (XO: 1.3497676)

 - Block number: 2 starting at 2018-07-31T03:01:08.982.135.750 (XO: 15.9822782, t=35.6°C)  Rank: 00004
   TM values: Max =     7  Min =   -15  Ave =    -4.42  Std =     2.67
   Min TM value:-15 at an offset of 127.625 usec -> 2018-07-31T03:01:08.982.263.375 (XO: 15.9824059)
   Max TM value:7 at an offset of 128.963 usec -> 2018-07-31T03:01:08.982.264.713 (XO: 15.9824072)
   Max dTM/dt [1/us]::6.90e+01 at an offset of 288.526 usec -> 2018-07-31T03:01:08.982.424.276 (XO: 15.9825668)
   Min dTM/dt [1/us]::-6.90e+01 at an offset of 129.301 usec -> 2018-07-31T03:01:08.982.265.050 (XO: 15.9824075)

 - Block number: 3 starting at 2018-07-31T03:01:25.490.360.750 (XO: 32.4905024, t=35.6°C)  Rank: 00004
   TM values: Max =     7  Min =   -16  Ave =    -4.38  Std =     2.73
   Min TM value:-16 at an offset of 133.776 usec -> 2018-07-31T03:01:25.490.494.525 (XO: 32.4906362)
   Max TM value:7 at an offset of 122.713 usec -> 2018-07-31T03:01:25.490.483.462 (XO: 32.4906251)
   Max dTM/dt [1/us]::7.50e+01 at an offset of 187.251 usec -> 2018-07-31T03:01:25.490.548.000 (XO: 32.4906897)
   Min dTM/dt [1/us]::-7.30e+01 at an offset of 262.689 usec -> 2018-07-31T03:01:25.490.623.438 (XO: 32.4907651)
