-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180725_234021_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.541147 to 34.281349
-each separately from 8s to 34s and from 87390us to 541436us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.545954 in 1 packets - #/No: 18/18
      17.193381 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-25T23:40:29.541.147.000 to 2018-07-25T23:40:55.281.349.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.545954 in 1 packets - #/No: 18/18
      17.193381 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4751476 to 35.4750415
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999633 in 1 packets - #/No: 18/18
      16.9999306 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180725_234021_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.0162938 to 10.0165828
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.5459190  -- OK for SL --
   MIU time from 2018-07-25T23:40:29.541.147.000 to 2018-07-25T23:40:29.541.436.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.545954

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.5625018 to 18.5627908
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1933110  -- OK for SL --
   MIU time from 2018-07-25T23:40:38.087.390.000 to 2018-07-25T23:40:38.087.679.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.193381

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.7561018 to 35.7563908
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-25T23:40:55.281.060.000 to 2018-07-25T23:40:55.281.349.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-25T23:40:29.541.006.750
   Max SL corrected packet time: 2018-07-25T23:40:55.281.208.750
   Min SL corrected XO rel time: 8.5411484
   Max SL corrected XO rel time: 34.2813505
   TM values from -19 to 11 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-25T23:40:29.541.006.750 (XO: 8.5411484, t=35.6°C)  Rank: 00005
   TM values: Max =    11  Min =   -17  Ave =    -4.31  Std =     2.96
   Min TM value:-17 at an offset of 96.675 usec -> 2018-07-25T23:40:29.541.103.425 (XO: 8.5412450)
   Max TM value:11 at an offset of 108.388 usec -> 2018-07-25T23:40:29.541.115.137 (XO: 8.5412568)
   Max dTM/dt [1/us]::1.06e+02 at an offset of 30.000 usec -> 2018-07-25T23:40:29.541.036.750 (XO: 8.5411784)
   Min dTM/dt [1/us]::-1.09e+02 at an offset of 79.300 usec -> 2018-07-25T23:40:29.541.086.050 (XO: 8.5412277)

 - Block number: 2 starting at 2018-07-25T23:40:38.087.249.750 (XO: 17.0873913, t=35.6°C)  Rank: 00004
   TM values: Max =    10  Min =   -19  Ave =    -4.27  Std =     3.17
   Min TM value:-19 at an offset of 108.838 usec -> 2018-07-25T23:40:38.087.358.587 (XO: 17.0875001)
   Max TM value:10 at an offset of 107.225 usec -> 2018-07-25T23:40:38.087.356.975 (XO: 17.0874985)
   Max dTM/dt [1/us]::1.06e+02 at an offset of 234.676 usec -> 2018-07-25T23:40:38.087.484.425 (XO: 17.0876259)
   Min dTM/dt [1/us]::-1.19e+02 at an offset of 5.750 usec -> 2018-07-25T23:40:38.087.255.500 (XO: 17.0873970)

 - Block number: 3 starting at 2018-07-25T23:40:55.280.919.750 (XO: 34.2810615, t=35.6°C)  Rank: 00004
   TM values: Max =    10  Min =   -17  Ave =    -4.40  Std =     3.51
   Min TM value:-17 at an offset of 83.688 usec -> 2018-07-25T23:40:55.281.003.437 (XO: 34.2811451)
   Max TM value:10 at an offset of 82.013 usec -> 2018-07-25T23:40:55.281.001.762 (XO: 34.2811435)
   Max dTM/dt [1/us]::1.36e+02 at an offset of 225.488 usec -> 2018-07-25T23:40:55.281.145.238 (XO: 34.2812869)
   Min dTM/dt [1/us]::-1.32e+02 at an offset of 225.563 usec -> 2018-07-25T23:40:55.281.145.313 (XO: 34.2812870)
