-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180724_235738_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.406987 to 35.612191
-each separately from 0s to 35s and from 406987us to 683836us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.276271 in 1 packets - #/No: 18/18
      17.928066 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-24T23:57:38.406.987.000 to 2018-07-24T23:58:13.612.191.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.276271 in 1 packets - #/No: 18/18
      17.928066 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4362560 to 36.4361252
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999364 in 1 packets - #/No: 18/18
      17.9999328 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180724_235738_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8432427 to 1.8435317
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.2762070  -- OK for SL --
   MIU time from 2018-07-24T23:57:38.406.987.000 to 2018-07-24T23:57:38.407.276.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.276271

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1197387 to 19.1200277
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.9279990  -- OK for SL --
   MIU time from 2018-07-24T23:57:55.683.547.000 to 2018-07-24T23:57:55.683.836.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.928066

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.0480267 to 37.0483157
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-24T23:58:13.611.902.000 to 2018-07-24T23:58:13.612.191.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-24T23:57:38.406.846.750
   Max SL corrected packet time: 2018-07-24T23:58:13.612.050.750
   Min SL corrected XO rel time: 0.4069882
   Max SL corrected XO rel time: 35.6121928
   TM values from -45 to 42 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-24T23:57:38.406.846.750 (XO: 0.4069882, t=35.6°C)  Rank: 00005
   TM values: Max =    11  Min =   -17  Ave =    -4.39  Std =     3.25
   Min TM value:-17 at an offset of 129.638 usec -> 2018-07-24T23:57:38.406.976.387 (XO: 0.4071179)
   Max TM value:11 at an offset of 127.700 usec -> 2018-07-24T23:57:38.406.974.450 (XO: 0.4071159)
   Max dTM/dt [1/us]::1.23e+02 at an offset of 285.351 usec -> 2018-07-24T23:57:38.407.132.101 (XO: 0.4072736)
   Min dTM/dt [1/us]::-1.23e+02 at an offset of 280.914 usec -> 2018-07-24T23:57:38.407.127.663 (XO: 0.4072692)

 - Block number: 2 starting at 2018-07-24T23:57:55.683.406.750 (XO: 17.6835488, t=35.6°C)  Rank: 00006
   TM values: Max =    15  Min =   -20  Ave =    -4.47  Std =     3.54
   Min TM value:-20 at an offset of 115.913 usec -> 2018-07-24T23:57:55.683.522.662 (XO: 17.6836647)
   Max TM value:15 at an offset of 120.625 usec -> 2018-07-24T23:57:55.683.527.375 (XO: 17.6836695)
   Max dTM/dt [1/us]::1.27e+02 at an offset of 158.938 usec -> 2018-07-24T23:57:55.683.565.688 (XO: 17.6837078)
   Min dTM/dt [1/us]::-1.11e+02 at an offset of 181.338 usec -> 2018-07-24T23:57:55.683.588.088 (XO: 17.6837302)

 - Block number: 3 starting at 2018-07-24T23:58:13.611.761.750 (XO: 35.6119038, t=35.6°C)  Rank: 00005
   TM values: Max =    42  Min =   -45  Ave =    -4.37  Std =     4.34
   Min TM value:-45 at an offset of 153.513 usec -> 2018-07-24T23:58:13.611.915.263 (XO: 35.6120573)
   Max TM value:42 at an offset of 68.213 usec -> 2018-07-24T23:58:13.611.829.962 (XO: 35.6119720)
   Max dTM/dt [1/us]::3.48e+02 at an offset of 153.613 usec -> 2018-07-24T23:58:13.611.915.363 (XO: 35.6120574)
   Min dTM/dt [1/us]::-3.29e+02 at an offset of 68.288 usec -> 2018-07-24T23:58:13.611.830.037 (XO: 35.6119721)
