-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180723_125710_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.051546 to 33.245314
-each separately from 0s to 33s and from 50887us to 245314us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.999052 in 1 packets - #/No: 18/18
      18.193849 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-23T12:57:10.051.546.000 to 2018-07-23T12:57:43.245.314.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.999052 in 1 packets - #/No: 18/18
      18.193849 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4343096 to 34.4341756
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999391 in 1 packets - #/No: 18/18
      17.9999269 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180723_125710_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4858569 to 1.4861459
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9989910  -- OK for SL --
   MIU time from 2018-07-23T12:57:10.051.546.000 to 2018-07-23T12:57:10.051.835.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.999052

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4851369 to 16.4854259
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.1937750  -- OK for SL --
   MIU time from 2018-07-23T12:57:25.050.887.000 to 2018-07-23T12:57:25.051.176.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.193849

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.6792009 to 34.6794899
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-23T12:57:43.245.025.000 to 2018-07-23T12:57:43.245.314.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-23T12:57:10.051.405.750
   Max SL corrected packet time: 2018-07-23T12:57:43.245.173.750
   Min SL corrected XO rel time: 0.0515475
   Max SL corrected XO rel time: 33.2453153
   TM values from -17 to 8 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-23T12:57:10.051.405.750 (XO: 0.0515475, t=37.2°C)  Rank: 00003
   TM values: Max =     6  Min =   -16  Ave =    -4.35  Std =     2.47
   Min TM value:-16 at an offset of 91.163 usec -> 2018-07-23T12:57:10.051.496.912 (XO: 0.0516387)
   Max TM value:6 at an offset of 85.838 usec -> 2018-07-23T12:57:10.051.491.587 (XO: 0.0516333)
   Max dTM/dt [1/us]::4.70e+01 at an offset of 85.088 usec -> 2018-07-23T12:57:10.051.490.837 (XO: 0.0516326)
   Min dTM/dt [1/us]::-5.30e+01 at an offset of 137.588 usec -> 2018-07-23T12:57:10.051.543.338 (XO: 0.0516851)

 - Block number: 2 starting at 2018-07-23T12:57:25.050.746.750 (XO: 15.0508884, t=37.2°C)  Rank: 00003
   TM values: Max =     8  Min =   -17  Ave =    -4.31  Std =     3.02
   Min TM value:-17 at an offset of 101.875 usec -> 2018-07-23T12:57:25.050.848.625 (XO: 15.0509903)
   Max TM value:8 at an offset of 120.850 usec -> 2018-07-23T12:57:25.050.867.600 (XO: 15.0510093)
   Max dTM/dt [1/us]::5.60e+01 at an offset of 192.026 usec -> 2018-07-23T12:57:25.050.938.775 (XO: 15.0510804)
   Min dTM/dt [1/us]::-6.00e+01 at an offset of 34.775 usec -> 2018-07-23T12:57:25.050.781.525 (XO: 15.0509232)

 - Block number: 3 starting at 2018-07-23T12:57:43.244.884.750 (XO: 33.2450263, t=37.2°C)  Rank: 00004
   TM values: Max =     8  Min =   -17  Ave =    -4.45  Std =     3.10
   Min TM value:-17 at an offset of 122.063 usec -> 2018-07-23T12:57:43.245.006.813 (XO: 33.2451484)
   Max TM value:8 at an offset of 252.414 usec -> 2018-07-23T12:57:43.245.137.163 (XO: 33.2452787)
   Max dTM/dt [1/us]::6.30e+01 at an offset of 206.401 usec -> 2018-07-23T12:57:43.245.091.150 (XO: 33.2452327)
   Min dTM/dt [1/us]::-5.60e+01 at an offset of 143.076 usec -> 2018-07-23T12:57:43.245.027.825 (XO: 33.2451694)
