-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180719_193040_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.321987 to 34.416704
-each separately from 3s to 34s and from 321987us to 454960us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.132395 in 1 packets - #/No: 18/18
      7.961455 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-19T19:30:44.321.987.000 to 2018-07-19T19:31:15.416.704.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.132395 in 1 packets - #/No: 18/18
      7.961455 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4285491 to 35.4284253
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.9999082 in 1 packets - #/No: 18/18
      7.9999680 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180719_193040_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.7505362 to 4.7508252
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.1323030  -- OK for SL --
   MIU time from 2018-07-19T19:30:44.321.987.000 to 2018-07-19T19:30:44.322.276.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.132395

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.8831282 to 27.8834172
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.9614230  -- OK for SL --
   MIU time from 2018-07-19T19:31:07.454.671.000 to 2018-07-19T19:31:07.454.960.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.961455

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.8448402 to 35.8451292
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-19T19:31:15.416.415.000 to 2018-07-19T19:31:15.416.704.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-19T19:30:44.321.846.750
   Max SL corrected packet time: 2018-07-19T19:31:15.416.563.750
   Min SL corrected XO rel time: 3.3219884
   Max SL corrected XO rel time: 34.4167055
   TM values from -42 to 34 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-19T19:30:44.321.846.750 (XO: 3.3219884, t=34.0°C)  Rank: 00007
   TM values: Max =    31  Min =   -42  Ave =    -4.38  Std =    10.74
   Min TM value:-42 at an offset of 150.338 usec -> 2018-07-19T19:30:44.321.997.088 (XO: 3.3221388)
   Max TM value:31 at an offset of 109.600 usec -> 2018-07-19T19:30:44.321.956.350 (XO: 3.3220980)
   Max dTM/dt [1/us]::2.95e+02 at an offset of 61.613 usec -> 2018-07-19T19:30:44.321.908.362 (XO: 3.3220500)
   Min dTM/dt [1/us]::-3.18e+02 at an offset of 26.525 usec -> 2018-07-19T19:30:44.321.873.275 (XO: 3.3220149)

 - Block number: 2 starting at 2018-07-19T19:31:07.454.530.750 (XO: 26.4546728, t=34.0°C)  Rank: 00008
   TM values: Max =    34  Min =   -41  Ave =    -4.51  Std =    10.18
   Min TM value:-41 at an offset of 201.338 usec -> 2018-07-19T19:31:07.454.732.088 (XO: 26.4548741)
   Max TM value:34 at an offset of 145.276 usec -> 2018-07-19T19:31:07.454.676.025 (XO: 26.4548180)
   Max dTM/dt [1/us]::4.50e+02 at an offset of 197.338 usec -> 2018-07-19T19:31:07.454.728.088 (XO: 26.4548701)
   Min dTM/dt [1/us]::-4.09e+02 at an offset of 115.513 usec -> 2018-07-19T19:31:07.454.646.262 (XO: 26.4547883)

 - Block number: 3 starting at 2018-07-19T19:31:15.416.274.750 (XO: 34.4164165, t=34.0°C)  Rank: 00006
   TM values: Max =    28  Min =   -35  Ave =    -4.49  Std =     8.41
   Min TM value:-35 at an offset of 133.288 usec -> 2018-07-19T19:31:15.416.408.038 (XO: 34.4165498)
   Max TM value:28 at an offset of 114.450 usec -> 2018-07-19T19:31:15.416.389.200 (XO: 34.4165310)
   Max dTM/dt [1/us]::4.14e+02 at an offset of 171.363 usec -> 2018-07-19T19:31:15.416.446.113 (XO: 34.4165879)
   Min dTM/dt [1/us]::-4.08e+02 at an offset of 122.375 usec -> 2018-07-19T19:31:15.416.397.125 (XO: 34.4165389)
