-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180717_001005_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.402646 to 35.840103
-each separately from 6s to 35s and from 339667us to 840103us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.936732 in 1 packets - #/No: 18/18
      13.499858 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-17T00:10:12.402.646.000 to 2018-07-17T00:10:41.840.103.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.936732 in 1 packets - #/No: 18/18
      13.499858 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4399762 to 36.4398663
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999393 in 1 packets - #/No: 18/18
      12.9999508 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180717_001005_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8426225 to 7.8429115
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9366710  -- OK for SL --
   MIU time from 2018-07-17T00:10:12.402.646.000 to 2018-07-17T00:10:12.402.935.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.936732

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.7795825 to 23.7798715
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4998070  -- OK for SL --
   MIU time from 2018-07-17T00:10:28.339.667.000 to 2018-07-17T00:10:28.339.956.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.499858

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.2796785 to 37.2799675
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-17T00:10:41.839.814.000 to 2018-07-17T00:10:41.840.103.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-17T00:10:12.402.505.750
   Max SL corrected packet time: 2018-07-17T00:10:41.839.962.750
   Min SL corrected XO rel time: 6.4026479
   Max SL corrected XO rel time: 35.8401044
   TM values from -33 to 27 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-17T00:10:12.402.505.750 (XO: 6.4026479, t=34.0°C)  Rank: 00006
   TM values: Max =    14  Min =   -22  Ave =    -4.39  Std =     6.37
   Min TM value:-22 at an offset of 89.550 usec -> 2018-07-17T00:10:12.402.595.300 (XO: 6.4027374)
   Max TM value:14 at an offset of 142.263 usec -> 2018-07-17T00:10:12.402.648.013 (XO: 6.4027901)
   Max dTM/dt [1/us]::2.26e+02 at an offset of 91.250 usec -> 2018-07-17T00:10:12.402.597.000 (XO: 6.4027391)
   Min dTM/dt [1/us]::-2.29e+02 at an offset of 100.800 usec -> 2018-07-17T00:10:12.402.606.550 (XO: 6.4027487)

 - Block number: 2 starting at 2018-07-17T00:10:28.339.526.750 (XO: 22.3396683, t=34.0°C)  Rank: 00004
   TM values: Max =    14  Min =   -24  Ave =    -4.26  Std =     3.64
   Min TM value:-24 at an offset of 113.963 usec -> 2018-07-17T00:10:28.339.640.712 (XO: 22.3397822)
   Max TM value:14 at an offset of 122.275 usec -> 2018-07-17T00:10:28.339.649.025 (XO: 22.3397905)
   Max dTM/dt [1/us]::2.58e+02 at an offset of 119.300 usec -> 2018-07-17T00:10:28.339.646.050 (XO: 22.3397876)
   Min dTM/dt [1/us]::-2.43e+02 at an offset of 128.363 usec -> 2018-07-17T00:10:28.339.655.112 (XO: 22.3397966)

 - Block number: 3 starting at 2018-07-17T00:10:41.839.673.750 (XO: 35.8398154, t=34.0°C)  Rank: 00013
   TM values: Max =    27  Min =   -33  Ave =    -4.37  Std =    10.83
   Min TM value:-33 at an offset of 121.088 usec -> 2018-07-17T00:10:41.839.794.837 (XO: 35.8399365)
   Max TM value:27 at an offset of 128.350 usec -> 2018-07-17T00:10:41.839.802.100 (XO: 35.8399437)
   Max dTM/dt [1/us]::3.77e+02 at an offset of 126.963 usec -> 2018-07-17T00:10:41.839.800.712 (XO: 35.8399424)
   Min dTM/dt [1/us]::-3.69e+02 at an offset of 121.063 usec -> 2018-07-17T00:10:41.839.794.812 (XO: 35.8399365)
