-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180715_001821_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.302103 to 34.178388
-each separately from 0s to 34s and from 37924us to 302392us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.735532 in 1 packets - #/No: 18/18
      19.139886 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-15T00:18:23.302.103.000 to 2018-07-15T00:18:57.178.388.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.735532 in 1 packets - #/No: 18/18
      19.139886 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4386360 to 35.4384953
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999380 in 1 packets - #/No: 18/18
      18.9999213 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180715_001821_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7407393 to 1.7410283
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7354710  -- OK for SL --
   MIU time from 2018-07-15T00:18:23.302.103.000 to 2018-07-15T00:18:23.302.392.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.735532

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4764993 to 16.4767883
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1398070  -- OK for SL --
   MIU time from 2018-07-15T00:18:38.037.924.000 to 2018-07-15T00:18:38.038.213.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.139886

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.6165953 to 35.6168843
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-15T00:18:57.178.099.000 to 2018-07-15T00:18:57.178.388.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-15T00:18:23.301.962.750
   Max SL corrected packet time: 2018-07-15T00:18:57.178.247.750
   Min SL corrected XO rel time: 0.3021045
   Max SL corrected XO rel time: 34.1783897
   TM values from -52 to 55 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-15T00:18:23.301.962.750 (XO: 0.3021045, t=34.0°C)  Rank: 00007
   TM values: Max =    55  Min =   -52  Ave =    -4.33  Std =     8.00
   Min TM value:-52 at an offset of 128.226 usec -> 2018-07-15T00:18:23.302.090.975 (XO: 0.3022328)
   Max TM value:55 at an offset of 127.751 usec -> 2018-07-15T00:18:23.302.090.500 (XO: 0.3022323)
   Max dTM/dt [1/us]::9.37e+02 at an offset of 127.713 usec -> 2018-07-15T00:18:23.302.090.463 (XO: 0.3022323)
   Min dTM/dt [1/us]::-9.20e+02 at an offset of 128.188 usec -> 2018-07-15T00:18:23.302.090.938 (XO: 0.3022327)

 - Block number: 2 starting at 2018-07-15T00:18:38.037.783.750 (XO: 15.0379255, t=34.0°C)  Rank: 00006
   TM values: Max =    16  Min =   -25  Ave =    -4.27  Std =     6.79
   Min TM value:-25 at an offset of 82.738 usec -> 2018-07-15T00:18:38.037.866.487 (XO: 15.0380082)
   Max TM value:16 at an offset of 168.688 usec -> 2018-07-15T00:18:38.037.952.438 (XO: 15.0380941)
   Max dTM/dt [1/us]::2.70e+02 at an offset of 156.538 usec -> 2018-07-15T00:18:38.037.940.288 (XO: 15.0380820)
   Min dTM/dt [1/us]::-2.87e+02 at an offset of 136.838 usec -> 2018-07-15T00:18:38.037.920.588 (XO: 15.0380623)

 - Block number: 3 starting at 2018-07-15T00:18:57.177.958.750 (XO: 34.1781007, t=34.0°C)  Rank: 00009
   TM values: Max =    23  Min =   -29  Ave =    -4.28  Std =     8.93
   Min TM value:-29 at an offset of 130.313 usec -> 2018-07-15T00:18:57.178.089.063 (XO: 34.1782311)
   Max TM value:23 at an offset of 176.551 usec -> 2018-07-15T00:18:57.178.135.300 (XO: 34.1782773)
   Max dTM/dt [1/us]::3.31e+02 at an offset of 176.526 usec -> 2018-07-15T00:18:57.178.135.275 (XO: 34.1782773)
   Min dTM/dt [1/us]::-3.25e+02 at an offset of 153.701 usec -> 2018-07-15T00:18:57.178.112.450 (XO: 34.1782544)
