-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180710_181717_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.413629 to 30.547259
-each separately from 14s to 30s and from 413629us to 560149us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.145942 in 1 packets - #/No: 18/18
      9.986821 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-10T18:17:33.413.629.000 to 2018-07-10T18:17:49.547.259.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.145942 in 1 packets - #/No: 18/18
      9.986821 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4380541 to 31.4379927
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.9999769 in 1 packets - #/No: 18/18
      9.9999617 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180710_181717_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      19 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.8516827 to 15.8519717
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.1459190  -- OK for SL --
   MIU time from 2018-07-10T18:17:33.413.629.000 to 2018-07-10T18:17:33.413.918.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.145942

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      25 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.9978907 to 21.9981797
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9867830  -- OK for SL --
   MIU time from 2018-07-10T18:17:39.559.860.000 to 2018-07-10T18:17:39.560.149.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.986821

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      16 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9849627 to 31.9852517
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-10T18:17:49.546.970.000 to 2018-07-10T18:17:49.547.259.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-10T18:17:33.413.488.750
   Max SL corrected packet time: 2018-07-10T18:17:49.547.118.750
   Min SL corrected XO rel time: 14.4136302
   Max SL corrected XO rel time: 30.5472611
   TM values from -82 to 74 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-10T18:17:33.413.488.750 (XO: 14.4136302, t=35.6°C)  Rank: 00019
   TM values: Max =    72  Min =   -74  Ave =    -4.37  Std =    27.12
   Min TM value:-74 at an offset of 154.376 usec -> 2018-07-10T18:17:33.413.643.125 (XO: 14.4137846)
   Max TM value:72 at an offset of 154.413 usec -> 2018-07-10T18:17:33.413.643.163 (XO: 14.4137846)
   Max dTM/dt [1/us]::6.17e+02 at an offset of 116.513 usec -> 2018-07-10T18:17:33.413.605.262 (XO: 14.4137467)
   Min dTM/dt [1/us]::-6.41e+02 at an offset of 216.288 usec -> 2018-07-10T18:17:33.413.705.038 (XO: 14.4138465)

 - Block number: 2 starting at 2018-07-10T18:17:39.559.719.750 (XO: 20.5598618, t=35.6°C)  Rank: 00025
   TM values: Max =    74  Min =   -82  Ave =    -4.32  Std =    27.80
   Min TM value:-82 at an offset of 147.826 usec -> 2018-07-10T18:17:39.559.867.575 (XO: 20.5600097)
   Max TM value:74 at an offset of 147.026 usec -> 2018-07-10T18:17:39.559.866.775 (XO: 20.5600089)
   Max dTM/dt [1/us]::7.00e+02 at an offset of 147.426 usec -> 2018-07-10T18:17:39.559.867.175 (XO: 20.5600093)
   Min dTM/dt [1/us]::-7.31e+02 at an offset of 156.026 usec -> 2018-07-10T18:17:39.559.875.775 (XO: 20.5600179)

 - Block number: 3 starting at 2018-07-10T18:17:49.546.829.750 (XO: 30.5469721, t=35.6°C)  Rank: 00016
   TM values: Max =    73  Min =   -82  Ave =    -4.33  Std =    19.82
   Min TM value:-82 at an offset of 140.051 usec -> 2018-07-10T18:17:49.546.969.800 (XO: 30.5471121)
   Max TM value:73 at an offset of 101.263 usec -> 2018-07-10T18:17:49.546.931.012 (XO: 30.5470733)
   Max dTM/dt [1/us]::6.06e+02 at an offset of 140.088 usec -> 2018-07-10T18:17:49.546.969.838 (XO: 30.5471122)
   Min dTM/dt [1/us]::-5.87e+02 at an offset of 162.626 usec -> 2018-07-10T18:17:49.546.992.375 (XO: 30.5471347)
