-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180702_005609_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.596183 to 41.698104
-each separately from 2s to 41s and from 213300us to 698104us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.616828 in 1 packets - #/No: 18/18
      16.484226 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-02T00:56:14.596.183.000 to 2018-07-02T00:56:53.698.104.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.616828 in 1 packets - #/No: 18/18
      16.484226 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4377008 to 42.4375407
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.9999056 in 1 packets - #/No: 18/18
      15.9999343 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180702_005609_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0338825 to 4.0341715
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6167350  -- OK for SL --
   MIU time from 2018-07-02T00:56:14.596.183.000 to 2018-07-02T00:56:14.596.472.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.616828

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6509065 to 26.6511955
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4841590  -- OK for SL --
   MIU time from 2018-07-02T00:56:37.213.300.000 to 2018-07-02T00:56:37.213.589.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.484226

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.1353545 to 43.1356435
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-02T00:56:53.697.815.000 to 2018-07-02T00:56:53.698.104.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-02T00:56:14.596.042.750
   Max SL corrected packet time: 2018-07-02T00:56:53.697.963.750
   Min SL corrected XO rel time: 2.5961841
   Max SL corrected XO rel time: 41.6981057
   TM values from -23 to 13 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-02T00:56:14.596.042.750 (XO: 2.5961841, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -19  Ave =    -4.32  Std =     4.24
   Min TM value:-19 at an offset of 116.900 usec -> 2018-07-02T00:56:14.596.159.650 (XO: 2.5963010)
   Max TM value:10 at an offset of 116.325 usec -> 2018-07-02T00:56:14.596.159.075 (XO: 2.5963005)
   Max dTM/dt [1/us]::1.93e+02 at an offset of 117.125 usec -> 2018-07-02T00:56:14.596.159.875 (XO: 2.5963013)
   Min dTM/dt [1/us]::-2.02e+02 at an offset of 117.700 usec -> 2018-07-02T00:56:14.596.160.450 (XO: 2.5963018)

 - Block number: 2 starting at 2018-07-02T00:56:37.213.159.750 (XO: 25.2133010, t=34.0°C)  Rank: 00003
   TM values: Max =    13  Min =   -23  Ave =    -4.17  Std =     5.70
   Min TM value:-23 at an offset of 180.776 usec -> 2018-07-02T00:56:37.213.340.525 (XO: 25.2134818)
   Max TM value:13 at an offset of 29.525 usec -> 2018-07-02T00:56:37.213.189.275 (XO: 25.2133305)
   Max dTM/dt [1/us]::2.58e+02 at an offset of 128.988 usec -> 2018-07-02T00:56:37.213.288.738 (XO: 25.2134300)
   Min dTM/dt [1/us]::-2.62e+02 at an offset of 169.263 usec -> 2018-07-02T00:56:37.213.329.013 (XO: 25.2134702)

 - Block number: 3 starting at 2018-07-02T00:56:53.697.674.750 (XO: 41.6978167, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -19  Ave =    -4.37  Std =     4.58
   Min TM value:-19 at an offset of 105.988 usec -> 2018-07-02T00:56:53.697.780.737 (XO: 41.6979226)
   Max TM value:11 at an offset of 305.089 usec -> 2018-07-02T00:56:53.697.979.838 (XO: 41.6981217)
   Max dTM/dt [1/us]::2.66e+02 at an offset of 230.788 usec -> 2018-07-02T00:56:53.697.905.538 (XO: 41.6980474)
   Min dTM/dt [1/us]::-2.27e+02 at an offset of 305.126 usec -> 2018-07-02T00:56:53.697.979.876 (XO: 41.6981218)
