-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180629_093623_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.123934 to 30.077975
-each separately from 0s to 30s and from 41050us to 124223us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.916827 in 1 packets - #/No: 18/18
      15.036347 in 1 packets - #/No: 36/18
MIU packet time from 2018-06-29T09:36:26.123.934.000 to 2018-06-29T09:36:56.077.975.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.916827 in 1 packets - #/No: 18/18
      15.036347 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4410125 to 31.4408926
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999400 in 1 packets - #/No: 18/18
      14.9999401 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180629_093623_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5649475 to 1.5652365
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9167670  -- OK for SL --
   MIU time from 2018-06-29T09:36:26.123.934.000 to 2018-06-29T09:36:26.124.223.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.916827

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4820035 to 16.4822925
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0362870  -- OK for SL --
   MIU time from 2018-06-29T09:36:41.041.050.000 to 2018-06-29T09:36:41.041.339.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.036347

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5185795 to 31.5188685
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-06-29T09:36:56.077.686.000 to 2018-06-29T09:36:56.077.975.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-06-29T09:36:26.123.793.750
   Max SL corrected packet time: 2018-06-29T09:36:56.077.834.750
   Min SL corrected XO rel time: 0.1239355
   Max SL corrected XO rel time: 30.0779762
   TM values from -13 to 4 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-06-29T09:36:26.123.793.750 (XO: 0.1239355, t=34.0°C)  Rank: 00002
   TM values: Max =     4  Min =   -12  Ave =    -4.37  Std =     3.85
   Min TM value:-12 at an offset of 8.213 usec -> 2018-06-29T09:36:26.123.801.962 (XO: 0.1239437)
   Max TM value:4 at an offset of 77.600 usec -> 2018-06-29T09:36:26.123.871.350 (XO: 0.1240131)
   Max dTM/dt [1/us]::5.10e+01 at an offset of 36.238 usec -> 2018-06-29T09:36:26.123.829.987 (XO: 0.1239717)
   Min dTM/dt [1/us]::-5.80e+01 at an offset of 198.013 usec -> 2018-06-29T09:36:26.123.991.763 (XO: 0.1241335)

 - Block number: 2 starting at 2018-06-29T09:36:41.040.909.750 (XO: 15.0410512, t=34.0°C)  Rank: 00002
   TM values: Max =     3  Min =   -12  Ave =    -4.48  Std =     3.04
   Min TM value:-12 at an offset of 49.400 usec -> 2018-06-29T09:36:41.040.959.150 (XO: 15.0411006)
   Max TM value:3 at an offset of 50.213 usec -> 2018-06-29T09:36:41.040.959.962 (XO: 15.0411014)
   Max dTM/dt [1/us]::5.00e+01 at an offset of 101.550 usec -> 2018-06-29T09:36:41.041.011.300 (XO: 15.0411527)
   Min dTM/dt [1/us]::-5.70e+01 at an offset of 110.038 usec -> 2018-06-29T09:36:41.041.019.787 (XO: 15.0411612)

 - Block number: 3 starting at 2018-06-29T09:36:56.077.545.750 (XO: 30.0776872, t=34.0°C)  Rank: 00002
   TM values: Max =     4  Min =   -13  Ave =    -4.34  Std =     3.73
   Min TM value:-13 at an offset of 87.750 usec -> 2018-06-29T09:36:56.077.633.500 (XO: 30.0777750)
   Max TM value:4 at an offset of 85.363 usec -> 2018-06-29T09:36:56.077.631.112 (XO: 30.0777726)
   Max dTM/dt [1/us]::4.70e+01 at an offset of 62.238 usec -> 2018-06-29T09:36:56.077.607.987 (XO: 30.0777494)
   Min dTM/dt [1/us]::-4.70e+01 at an offset of 163.338 usec -> 2018-06-29T09:36:56.077.709.088 (XO: 30.0778505)
