-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180628_092255_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.349944 to 32.681603
-each separately from 14s to 32s and from 251196us to 681603us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.900963 in 1 packets - #/No: 18/18
      17.429829 in 1 packets - #/No: 36/18
MIU packet time from 2018-06-28T09:23:12.349.944.000 to 2018-06-28T09:23:30.681.603.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.900963 in 1 packets - #/No: 18/18
      17.429829 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4317939 to 33.4317218
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.9999960 in 1 packets - #/No: 18/18
      16.9999319 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180628_092255_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.7817384 to 15.7820274
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.9009590  -- OK for SL --
   MIU time from 2018-06-28T09:23:12.349.944.000 to 2018-06-28T09:23:12.350.233.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.900963

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6829864 to 16.6832754
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4297590  -- OK for SL --
   MIU time from 2018-06-28T09:23:13.251.196.000 to 2018-06-28T09:23:13.251.485.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.429829

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.1130344 to 34.1133234
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-06-28T09:23:30.681.314.000 to 2018-06-28T09:23:30.681.603.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-06-28T09:23:12.349.803.750
   Max SL corrected packet time: 2018-06-28T09:23:30.681.462.750
   Min SL corrected XO rel time: 14.3499459
   Max SL corrected XO rel time: 32.6816043
   TM values from -14 to 5 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-06-28T09:23:12.349.803.750 (XO: 14.3499459, t=34.0°C)  Rank: 00002
   TM values: Max =     4  Min =   -14  Ave =    -4.38  Std =     3.32
   Min TM value:-14 at an offset of 247.526 usec -> 2018-06-28T09:23:12.350.051.275 (XO: 14.3501934)
   Max TM value:4 at an offset of 73.325 usec -> 2018-06-28T09:23:12.349.877.075 (XO: 14.3500192)
   Max dTM/dt [1/us]::6.70e+01 at an offset of 112.525 usec -> 2018-06-28T09:23:12.349.916.275 (XO: 14.3500584)
   Min dTM/dt [1/us]::-6.50e+01 at an offset of 148.326 usec -> 2018-06-28T09:23:12.349.952.075 (XO: 14.3500942)

 - Block number: 2 starting at 2018-06-28T09:23:13.251.055.750 (XO: 15.2511975, t=34.0°C)  Rank: 00002
   TM values: Max =     5  Min =   -14  Ave =    -4.31  Std =     3.20
   Min TM value:-14 at an offset of 111.475 usec -> 2018-06-28T09:23:13.251.167.225 (XO: 15.2513090)
   Max TM value:5 at an offset of 81.950 usec -> 2018-06-28T09:23:13.251.137.700 (XO: 15.2512795)
   Max dTM/dt [1/us]::9.00e+01 at an offset of 79.938 usec -> 2018-06-28T09:23:13.251.135.687 (XO: 15.2512774)
   Min dTM/dt [1/us]::-7.40e+01 at an offset of 79.763 usec -> 2018-06-28T09:23:13.251.135.512 (XO: 15.2512773)

 - Block number: 3 starting at 2018-06-28T09:23:30.681.173.750 (XO: 32.6813153, t=34.0°C)  Rank: 00002
   TM values: Max =     5  Min =   -14  Ave =    -4.33  Std =     3.50
   Min TM value:-14 at an offset of 88.200 usec -> 2018-06-28T09:23:30.681.261.950 (XO: 32.6814035)
   Max TM value:5 at an offset of 98.113 usec -> 2018-06-28T09:23:30.681.271.862 (XO: 32.6814134)
   Max dTM/dt [1/us]::7.20e+01 at an offset of 62.413 usec -> 2018-06-28T09:23:30.681.236.162 (XO: 32.6813777)
   Min dTM/dt [1/us]::-6.50e+01 at an offset of 19.613 usec -> 2018-06-28T09:23:30.681.193.362 (XO: 32.6813349)
