-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180626_072908_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.127833 to 30.288264
-each separately from 7s to 30s and from 127833us to 288264us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.042805 in 1 packets - #/No: 18/18
      2.116759 in 1 packets - #/No: 36/18
MIU packet time from 2018-06-26T07:29:18.127.833.000 to 2018-06-26T07:29:41.288.264.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.042805 in 1 packets - #/No: 18/18
      2.116759 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4404366 to 31.4403434
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999149 in 1 packets - #/No: 18/18
      1.9999919 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180626_072908_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.5682707 to 8.5685597
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0427190  -- OK for SL --
   MIU time from 2018-06-26T07:29:18.127.833.000 to 2018-06-26T07:29:18.128.122.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.042805

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      15 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.6112787 to 29.6115677
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.1167510  -- OK for SL --
   MIU time from 2018-06-26T07:29:39.170.927.000 to 2018-06-26T07:29:39.171.216.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.116759

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7283187 to 31.7286077
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-06-26T07:29:41.287.975.000 to 2018-06-26T07:29:41.288.264.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-06-26T07:29:18.127.692.750
   Max SL corrected packet time: 2018-06-26T07:29:41.288.123.750
   Min SL corrected XO rel time: 7.1278346
   Max SL corrected XO rel time: 30.2882655
   TM values from -45 to 35 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-06-26T07:29:18.127.692.750 (XO: 7.1278346, t=34.0°C)  Rank: 00010
   TM values: Max =    29  Min =   -35  Ave =    -4.38  Std =     8.54
   Min TM value:-35 at an offset of 140.563 usec -> 2018-06-26T07:29:18.127.833.313 (XO: 7.1279752)
   Max TM value:29 at an offset of 141.838 usec -> 2018-06-26T07:29:18.127.834.588 (XO: 7.1279765)
   Max dTM/dt [1/us]::2.06e+02 at an offset of 130.751 usec -> 2018-06-26T07:29:18.127.823.500 (XO: 7.1279654)
   Min dTM/dt [1/us]::-2.10e+02 at an offset of 142.476 usec -> 2018-06-26T07:29:18.127.835.225 (XO: 7.1279771)

 - Block number: 2 starting at 2018-06-26T07:29:39.170.786.750 (XO: 28.1709279, t=34.0°C)  Rank: 00015
   TM values: Max =    35  Min =   -45  Ave =    -4.36  Std =    14.29
   Min TM value:-45 at an offset of 175.838 usec -> 2018-06-26T07:29:39.170.962.588 (XO: 28.1711037)
   Max TM value:35 at an offset of 175.151 usec -> 2018-06-26T07:29:39.170.961.900 (XO: 28.1711031)
   Max dTM/dt [1/us]::3.14e+02 at an offset of 305.989 usec -> 2018-06-26T07:29:39.171.092.738 (XO: 28.1712339)
   Min dTM/dt [1/us]::-2.97e+02 at an offset of 175.788 usec -> 2018-06-26T07:29:39.170.962.538 (XO: 28.1711037)

 - Block number: 3 starting at 2018-06-26T07:29:41.287.834.750 (XO: 30.2879765, t=34.0°C)  Rank: 00013
   TM values: Max =    31  Min =   -42  Ave =    -4.42  Std =    12.96
   Min TM value:-42 at an offset of 125.276 usec -> 2018-06-26T07:29:41.287.960.025 (XO: 30.2881018)
   Max TM value:31 at an offset of 124.651 usec -> 2018-06-26T07:29:41.287.959.400 (XO: 30.2881011)
   Max dTM/dt [1/us]::2.71e+02 at an offset of 124.601 usec -> 2018-06-26T07:29:41.287.959.350 (XO: 30.2881011)
   Min dTM/dt [1/us]::-2.67e+02 at an offset of 125.288 usec -> 2018-06-26T07:29:41.287.960.038 (XO: 30.2881018)
