-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180621_225014_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.354810 to 34.710372
-each separately from 4s to 34s and from 354810us to 710372us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.169283 in 1 packets - #/No: 18/18
      17.185412 in 1 packets - #/No: 36/18
MIU packet time from 2018-06-21T22:50:21.354.810.000 to 2018-06-21T22:50:51.710.372.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.169283 in 1 packets - #/No: 18/18
      17.185412 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.4359072 to 35.4357878
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.9999483 in 1 packets - #/No: 18/18
      16.9999323 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180621_225014_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.7907173 to 5.7910063
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1692310  -- OK for SL --
   MIU time from 2018-06-21T22:50:21.354.810.000 to 2018-06-21T22:50:21.355.099.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.169283

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      18 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.9602373 to 18.9605263
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1853430  -- OK for SL --
   MIU time from 2018-06-21T22:50:34.524.382.000 to 2018-06-21T22:50:34.524.671.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.185412

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.1458693 to 36.1461583
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-06-21T22:50:51.710.083.000 to 2018-06-21T22:50:51.710.372.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-06-21T22:50:21.354.669.750
   Max SL corrected packet time: 2018-06-21T22:50:51.710.231.750
   Min SL corrected XO rel time: 4.3548115
   Max SL corrected XO rel time: 34.7103733
   TM values from -45 to 37 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-06-21T22:50:21.354.669.750 (XO: 4.3548115, t=34.0°C)  Rank: 00013
   TM values: Max =    29  Min =   -37  Ave =    -4.31  Std =     9.07
   Min TM value:-37 at an offset of 125.326 usec -> 2018-06-21T22:50:21.354.795.075 (XO: 4.3549368)
   Max TM value:29 at an offset of 124.425 usec -> 2018-06-21T22:50:21.354.794.175 (XO: 4.3549359)
   Max dTM/dt [1/us]::5.63e+02 at an offset of 124.388 usec -> 2018-06-21T22:50:21.354.794.137 (XO: 4.3549359)
   Min dTM/dt [1/us]::-5.74e+02 at an offset of 125.288 usec -> 2018-06-21T22:50:21.354.795.038 (XO: 4.3549368)

 - Block number: 2 starting at 2018-06-21T22:50:34.524.241.750 (XO: 17.5243839, t=34.0°C)  Rank: 00018
   TM values: Max =    37  Min =   -45  Ave =    -4.36  Std =    14.17
   Min TM value:-45 at an offset of 156.776 usec -> 2018-06-21T22:50:34.524.398.525 (XO: 17.5245407)
   Max TM value:37 at an offset of 123.976 usec -> 2018-06-21T22:50:34.524.365.725 (XO: 17.5245079)
   Max dTM/dt [1/us]::7.32e+02 at an offset of 123.938 usec -> 2018-06-21T22:50:34.524.365.688 (XO: 17.5245079)
   Min dTM/dt [1/us]::-7.05e+02 at an offset of 140.376 usec -> 2018-06-21T22:50:34.524.382.125 (XO: 17.5245243)

 - Block number: 3 starting at 2018-06-21T22:50:51.709.942.750 (XO: 34.7100843, t=34.0°C)  Rank: 00012
   TM values: Max =    23  Min =   -34  Ave =    -4.32  Std =    10.79
   Min TM value:-34 at an offset of 90.575 usec -> 2018-06-21T22:50:51.710.033.325 (XO: 34.7101749)
   Max TM value:23 at an offset of 90.225 usec -> 2018-06-21T22:50:51.710.032.975 (XO: 34.7101746)
   Max dTM/dt [1/us]::4.82e+02 at an offset of 111.275 usec -> 2018-06-21T22:50:51.710.054.025 (XO: 34.7101956)
   Min dTM/dt [1/us]::-4.81e+02 at an offset of 111.338 usec -> 2018-06-21T22:50:51.710.054.087 (XO: 34.7101957)
