-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180621_122138_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.115376 to 31.103640
-each separately from 1s to 31s and from 103351us to 649030us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.533076 in 1 packets - #/No: 18/18
      8.454321 in 1 packets - #/No: 36/18
MIU packet time from 2018-06-21T12:21:42.115.376.000 to 2018-06-21T12:22:12.103.640.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.533076 in 1 packets - #/No: 18/18
      8.454321 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4360162 to 32.4358971
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999166 in 1 packets - #/No: 18/18
      8.9999643 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180621_122138_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.5513931 to 2.5516821
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.5329910  -- OK for SL --
   MIU time from 2018-06-21T12:21:42.115.376.000 to 2018-06-21T12:21:42.115.665.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.533076

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.0846731 to 24.0849621
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.4542870  -- OK for SL --
   MIU time from 2018-06-21T12:22:03.648.741.000 to 2018-06-21T12:22:03.649.030.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.454321

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.5392491 to 32.5395381
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-06-21T12:22:12.103.351.000 to 2018-06-21T12:22:12.103.640.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-06-21T12:21:42.115.235.750
   Max SL corrected packet time: 2018-06-21T12:22:12.103.499.750
   Min SL corrected XO rel time: 1.1153773
   Max SL corrected XO rel time: 31.1036414
   TM values from -16 to 9 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-06-21T12:21:42.115.235.750 (XO: 1.1153773, t=34.0°C)  Rank: 00006
   TM values: Max =     9  Min =   -16  Ave =    -4.29  Std =     2.30
   Min TM value:-16 at an offset of 103.275 usec -> 2018-06-21T12:21:42.115.339.025 (XO: 1.1154806)
   Max TM value:9 at an offset of 99.425 usec -> 2018-06-21T12:21:42.115.335.175 (XO: 1.1154768)
   Max dTM/dt [1/us]::3.90e+01 at an offset of 258.989 usec -> 2018-06-21T12:21:42.115.494.738 (XO: 1.1156363)
   Min dTM/dt [1/us]::-4.70e+01 at an offset of 145.676 usec -> 2018-06-21T12:21:42.115.381.425 (XO: 1.1155230)

 - Block number: 2 starting at 2018-06-21T12:22:03.648.600.750 (XO: 22.6487428, t=34.0°C)  Rank: 00003
   TM values: Max =     3  Min =   -12  Ave =    -4.41  Std =     1.93
   Min TM value:-12 at an offset of 113.750 usec -> 2018-06-21T12:22:03.648.714.500 (XO: 22.6488566)
   Max TM value:3 at an offset of 102.588 usec -> 2018-06-21T12:22:03.648.703.337 (XO: 22.6488454)
   Max dTM/dt [1/us]::3.70e+01 at an offset of 170.376 usec -> 2018-06-21T12:22:03.648.771.125 (XO: 22.6489132)
   Min dTM/dt [1/us]::-4.90e+01 at an offset of 117.938 usec -> 2018-06-21T12:22:03.648.718.687 (XO: 22.6488608)

 - Block number: 3 starting at 2018-06-21T12:22:12.103.210.750 (XO: 31.1033524, t=34.0°C)  Rank: 00004
   TM values: Max =     5  Min =   -13  Ave =    -4.34  Std =     2.10
   Min TM value:-13 at an offset of 32.275 usec -> 2018-06-21T12:22:12.103.243.025 (XO: 31.1033847)
   Max TM value:5 at an offset of 78.900 usec -> 2018-06-21T12:22:12.103.289.650 (XO: 31.1034313)
   Max dTM/dt [1/us]::4.10e+01 at an offset of 287.026 usec -> 2018-06-21T12:22:12.103.497.776 (XO: 31.1036394)
   Min dTM/dt [1/us]::-4.10e+01 at an offset of 65.850 usec -> 2018-06-21T12:22:12.103.276.600 (XO: 31.1034183)
