-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180619_004955_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.770003 to 30.038277
-each separately from 1s to 30s and from 37988us to 770292us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.437989 in 1 packets - #/No: 18/18
      14.829418 in 1 packets - #/No: 36/18
MIU packet time from 2018-06-19T00:49:59.770.003.000 to 2018-06-19T00:50:28.038.277.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.437989 in 1 packets - #/No: 18/18
      14.829418 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4375443 to 31.4374287
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999444 in 1 packets - #/No: 18/18
      14.9999400 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180619_004955_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.2075458 to 3.2078348
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4379350  -- OK for SL --
   MIU time from 2018-06-19T00:49:59.770.003.000 to 2018-06-19T00:49:59.770.292.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.437989

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6457698 to 16.6460588
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8293590  -- OK for SL --
   MIU time from 2018-06-19T00:50:13.208.281.000 to 2018-06-19T00:50:13.208.570.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.829418

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4754178 to 31.4757068
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-06-19T00:50:28.037.988.000 to 2018-06-19T00:50:28.038.277.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-06-19T00:49:59.769.862.750
   Max SL corrected packet time: 2018-06-19T00:50:28.038.136.750
   Min SL corrected XO rel time: 1.7700046
   Max SL corrected XO rel time: 30.0382783
   TM values from -22 to 15 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-06-19T00:49:59.769.862.750 (XO: 1.7700046, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -21  Ave =    -4.43  Std =     5.99
   Min TM value:-21 at an offset of 101.813 usec -> 2018-06-19T00:49:59.769.964.562 (XO: 1.7701064)
   Max TM value:12 at an offset of 149.751 usec -> 2018-06-19T00:49:59.770.012.500 (XO: 1.7701543)
   Max dTM/dt [1/us]::2.66e+02 at an offset of 104.025 usec -> 2018-06-19T00:49:59.769.966.775 (XO: 1.7701086)
   Min dTM/dt [1/us]::-2.92e+02 at an offset of 182.188 usec -> 2018-06-19T00:49:59.770.044.938 (XO: 1.7701868)

 - Block number: 2 starting at 2018-06-19T00:50:13.208.140.750 (XO: 15.2082819, t=34.0°C)  Rank: 00003
   TM values: Max =    11  Min =   -20  Ave =    -4.42  Std =     4.49
   Min TM value:-20 at an offset of 123.950 usec -> 2018-06-19T00:50:13.208.264.700 (XO: 15.2084059)
   Max TM value:11 at an offset of 126.201 usec -> 2018-06-19T00:50:13.208.266.950 (XO: 15.2084081)
   Max dTM/dt [1/us]::2.55e+02 at an offset of 124.675 usec -> 2018-06-19T00:50:13.208.265.425 (XO: 15.2084066)
   Min dTM/dt [1/us]::-2.57e+02 at an offset of 123.913 usec -> 2018-06-19T00:50:13.208.264.662 (XO: 15.2084058)

 - Block number: 3 starting at 2018-06-19T00:50:28.037.847.750 (XO: 30.0379893, t=34.0°C)  Rank: 00003
   TM values: Max =    15  Min =   -22  Ave =    -4.28  Std =     6.85
   Min TM value:-22 at an offset of 100.238 usec -> 2018-06-19T00:50:28.037.947.987 (XO: 30.0380895)
   Max TM value:15 at an offset of 99.663 usec -> 2018-06-19T00:50:28.037.947.412 (XO: 30.0380889)
   Max dTM/dt [1/us]::2.96e+02 at an offset of 78.363 usec -> 2018-06-19T00:50:28.037.926.112 (XO: 30.0380676)
   Min dTM/dt [1/us]::-2.94e+02 at an offset of 108.000 usec -> 2018-06-19T00:50:28.037.955.750 (XO: 30.0380973)
