-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180608_194338_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.716345 to 30.296848
-each separately from 0s to 30s and from 296559us to 968586us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.251663 in 1 packets - #/No: 18/18
      1.327973 in 1 packets - #/No: 36/18
MIU packet time from 2018-06-08T19:43:41.716.345.000 to 2018-06-08T19:44:11.296.848.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.251663 in 1 packets - #/No: 18/18
      1.327973 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4368306 to 31.4367111
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      27.9998885 in 1 packets - #/No: 18/18
      1.9999920 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180608_194338_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      14 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1531742 to 2.1534632
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.2515510  -- OK for SL --
   MIU time from 2018-06-08T19:43:41.716.345.000 to 2018-06-08T19:43:41.716.634.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.251663

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      17 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.4050142 to 30.4053032
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.3279670  -- OK for SL --
   MIU time from 2018-06-08T19:44:09.968.297.000 to 2018-06-08T19:44:09.968.586.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.327973

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      18 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7332702 to 31.7335592
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-06-08T19:44:11.296.559.000 to 2018-06-08T19:44:11.296.848.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-06-08T19:43:41.716.204.750
   Max SL corrected packet time: 2018-06-08T19:44:11.296.707.750
   Min SL corrected XO rel time: 0.7163465
   Max SL corrected XO rel time: 30.2968493
   TM values from -48 to 38 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-06-08T19:43:41.716.204.750 (XO: 0.7163465, t=34.0°C)  Rank: 00014
   TM values: Max =    35  Min =   -41  Ave =    -4.46  Std =    11.74
   Min TM value:-41 at an offset of 96.138 usec -> 2018-06-08T19:43:41.716.300.887 (XO: 0.7164426)
   Max TM value:35 at an offset of 101.350 usec -> 2018-06-08T19:43:41.716.306.100 (XO: 0.7164478)
   Max dTM/dt [1/us]::3.09e+02 at an offset of 192.776 usec -> 2018-06-08T19:43:41.716.397.525 (XO: 0.7165392)
   Min dTM/dt [1/us]::-2.93e+02 at an offset of 191.876 usec -> 2018-06-08T19:43:41.716.396.625 (XO: 0.7165383)

 - Block number: 2 starting at 2018-06-08T19:44:09.968.156.750 (XO: 28.9682990, t=34.0°C)  Rank: 00017
   TM values: Max =    34  Min =   -45  Ave =    -4.51  Std =    14.37
   Min TM value:-45 at an offset of 125.463 usec -> 2018-06-08T19:44:09.968.282.213 (XO: 28.9684244)
   Max TM value:34 at an offset of 187.163 usec -> 2018-06-08T19:44:09.968.343.913 (XO: 28.9684861)
   Max dTM/dt [1/us]::3.19e+02 at an offset of 160.863 usec -> 2018-06-08T19:44:09.968.317.613 (XO: 28.9684598)
   Min dTM/dt [1/us]::-2.70e+02 at an offset of 160.113 usec -> 2018-06-08T19:44:09.968.316.863 (XO: 28.9684591)

 - Block number: 3 starting at 2018-06-08T19:44:11.296.418.750 (XO: 30.2965603, t=34.0°C)  Rank: 00018
   TM values: Max =    38  Min =   -48  Ave =    -4.53  Std =    13.31
   Min TM value:-48 at an offset of 78.500 usec -> 2018-06-08T19:44:11.296.497.250 (XO: 30.2966388)
   Max TM value:38 at an offset of 78.963 usec -> 2018-06-08T19:44:11.296.497.712 (XO: 30.2966392)
   Max dTM/dt [1/us]::3.16e+02 at an offset of 294.314 usec -> 2018-06-08T19:44:11.296.713.063 (XO: 30.2968546)
   Min dTM/dt [1/us]::-3.80e+02 at an offset of 292.889 usec -> 2018-06-08T19:44:11.296.711.638 (XO: 30.2968532)
