-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180608_184306_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.388978 to 30.069911
-each separately from 13s to 30s and from 69622us to 955478us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.565922 in 1 packets - #/No: 18/18
      4.114144 in 1 packets - #/No: 36/18
MIU packet time from 2018-06-08T18:43:22.388.978.000 to 2018-06-08T18:43:39.069.911.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.565922 in 1 packets - #/No: 18/18
      4.114144 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4356254 to 31.4355564
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999513 in 1 packets - #/No: 18/18
      4.9999797 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180608_184306_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      31 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.8246034 to 14.8248924
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5658710  -- OK for SL --
   MIU time from 2018-06-08T18:43:22.388.978.000 to 2018-06-08T18:43:22.389.267.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.565922

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      36 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.3907634 to 27.3910524
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.1141270  -- OK for SL --
   MIU time from 2018-06-08T18:43:34.955.189.000 to 2018-06-08T18:43:34.955.478.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.114144

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      28 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5051794 to 31.5054684
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-06-08T18:43:39.069.622.000 to 2018-06-08T18:43:39.069.911.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-06-08T18:43:22.388.837.750
   Max SL corrected packet time: 2018-06-08T18:43:39.069.770.750
   Min SL corrected XO rel time: 13.3889796
   Max SL corrected XO rel time: 30.0699123
   TM values from -116 to 105 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-06-08T18:43:22.388.837.750 (XO: 13.3889796, t=34.0°C)  Rank: 00031
   TM values: Max =    90  Min =   -98  Ave =    -4.39  Std =    29.26
   Min TM value:-98 at an offset of 138.263 usec -> 2018-06-08T18:43:22.388.976.013 (XO: 13.3891178)
   Max TM value:90 at an offset of 99.350 usec -> 2018-06-08T18:43:22.388.937.100 (XO: 13.3890789)
   Max dTM/dt [1/us]::8.63e+02 at an offset of 115.675 usec -> 2018-06-08T18:43:22.388.953.425 (XO: 13.3890952)
   Min dTM/dt [1/us]::-8.52e+02 at an offset of 115.638 usec -> 2018-06-08T18:43:22.388.953.387 (XO: 13.3890952)

 - Block number: 2 starting at 2018-06-08T18:43:34.955.048.750 (XO: 25.9551906, t=34.0°C)  Rank: 00036
   TM values: Max =   105  Min =  -116  Ave =    -4.31  Std =    38.83
   Min TM value:-116 at an offset of 115.388 usec -> 2018-06-08T18:43:34.955.164.137 (XO: 25.9553060)
   Max TM value:105 at an offset of 137.988 usec -> 2018-06-08T18:43:34.955.186.738 (XO: 25.9553286)
   Max dTM/dt [1/us]::9.82e+02 at an offset of 115.438 usec -> 2018-06-08T18:43:34.955.164.187 (XO: 25.9553061)
   Min dTM/dt [1/us]::-9.44e+02 at an offset of 138.426 usec -> 2018-06-08T18:43:34.955.187.175 (XO: 25.9553290)

 - Block number: 3 starting at 2018-06-08T18:43:39.069.481.750 (XO: 30.0696233, t=34.0°C)  Rank: 00028
   TM values: Max =    79  Min =  -100  Ave =    -4.42  Std =    30.71
   Min TM value:-100 at an offset of 144.501 usec -> 2018-06-08T18:43:39.069.626.250 (XO: 30.0697678)
   Max TM value:79 at an offset of 121.838 usec -> 2018-06-08T18:43:39.069.603.587 (XO: 30.0697451)
   Max dTM/dt [1/us]::8.56e+02 at an offset of 176.713 usec -> 2018-06-08T18:43:39.069.658.463 (XO: 30.0698000)
   Min dTM/dt [1/us]::-7.84e+02 at an offset of 115.175 usec -> 2018-06-08T18:43:39.069.596.925 (XO: 30.0697385)
