-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180528_095112_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.074328 to 33.743988
-each separately from 2s to 33s and from 11002us to 743988us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.936385 in 1 packets - #/No: 18/18
      18.732408 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-28T09:51:17.074.328.000 to 2018-05-28T09:51:48.743.988.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.936385 in 1 packets - #/No: 18/18
      18.732408 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4398091 to 34.4396882
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.9999492 in 1 packets - #/No: 18/18
      17.9999299 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180528_095112_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.5141376 to 3.5144266
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.9363350  -- OK for SL --
   MIU time from 2018-05-28T09:51:17.074.328.000 to 2018-05-28T09:51:17.074.617.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.936385

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4507616 to 16.4510506
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7323350  -- OK for SL --
   MIU time from 2018-05-28T09:51:30.011.002.000 to 2018-05-28T09:51:30.011.291.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.732408

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.1833856 to 35.1836746
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-28T09:51:48.743.699.000 to 2018-05-28T09:51:48.743.988.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-28T09:51:17.074.187.750
   Max SL corrected packet time: 2018-05-28T09:51:48.743.847.750
   Min SL corrected XO rel time: 2.0743288
   Max SL corrected XO rel time: 33.7439893
   TM values from -10 to 2 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-28T09:51:17.074.187.750 (XO: 2.0743288, t=34.0°C)  Rank: 00002
   TM values: Max =     0  Min =    -9  Ave =    -4.43  Std =     1.65
   Min TM value:-9 at an offset of 3.663 usec -> 2018-05-28T09:51:17.074.191.412 (XO: 2.0743325)
   Max TM value:0 at an offset of 19.763 usec -> 2018-05-28T09:51:17.074.207.512 (XO: 2.0743486)
   Max dTM/dt [1/us]::5.20e+01 at an offset of 87.125 usec -> 2018-05-28T09:51:17.074.274.875 (XO: 2.0744159)
   Min dTM/dt [1/us]::-4.30e+01 at an offset of 86.988 usec -> 2018-05-28T09:51:17.074.274.737 (XO: 2.0744158)

 - Block number: 2 starting at 2018-05-28T09:51:30.010.861.750 (XO: 15.0110033, t=34.0°C)  Rank: 00001
   TM values: Max =     2  Min =   -10  Ave =    -4.39  Std =     1.60
   Min TM value:-10 at an offset of 36.900 usec -> 2018-05-28T09:51:30.010.898.650 (XO: 15.0110402)
   Max TM value:2 at an offset of 200.588 usec -> 2018-05-28T09:51:30.011.062.338 (XO: 15.0112039)
   Max dTM/dt [1/us]::6.70e+01 at an offset of 200.476 usec -> 2018-05-28T09:51:30.011.062.225 (XO: 15.0112038)
   Min dTM/dt [1/us]::-7.20e+01 at an offset of 200.638 usec -> 2018-05-28T09:51:30.011.062.388 (XO: 15.0112040)

 - Block number: 3 starting at 2018-05-28T09:51:48.743.558.750 (XO: 33.7437003, t=34.0°C)  Rank: 00002
   TM values: Max =     1  Min =   -10  Ave =    -4.37  Std =     1.68
   Min TM value:-10 at an offset of 107.025 usec -> 2018-05-28T09:51:48.743.665.775 (XO: 33.7438073)
   Max TM value:1 at an offset of 31.750 usec -> 2018-05-28T09:51:48.743.590.500 (XO: 33.7437321)
   Max dTM/dt [1/us]::4.40e+01 at an offset of 189.113 usec -> 2018-05-28T09:51:48.743.747.863 (XO: 33.7438894)
   Min dTM/dt [1/us]::-5.80e+01 at an offset of 188.951 usec -> 2018-05-28T09:51:48.743.747.700 (XO: 33.7438893)
