-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180519_021405_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.473945 to 42.739163
-each separately from 0s to 42s and from 366852us to 739163us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.892618 in 1 packets - #/No: 18/18
      18.371733 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-19T02:14:08.473.945.000 to 2018-05-19T02:14:50.739.163.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.892618 in 1 packets - #/No: 18/18
      18.371733 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4415668 to 43.4414069
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999086 in 1 packets - #/No: 18/18
      17.9999315 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180519_021405_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9155114 to 1.9158004
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.8925270  -- OK for SL --
   MIU time from 2018-05-19T02:14:08.473.945.000 to 2018-05-19T02:14:08.474.234.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.892618

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.8083274 to 25.8086164
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3716630  -- OK for SL --
   MIU time from 2018-05-19T02:14:32.366.852.000 to 2018-05-19T02:14:32.367.141.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.371733

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.1802794 to 44.1805684
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-19T02:14:50.738.874.000 to 2018-05-19T02:14:50.739.163.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-19T02:14:08.473.804.750
   Max SL corrected packet time: 2018-05-19T02:14:50.739.022.750
   Min SL corrected XO rel time: 0.4739464
   Max SL corrected XO rel time: 42.7391643
   TM values from -26 to 17 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-19T02:14:08.473.804.750 (XO: 0.4739464, t=34.0°C)  Rank: 00007
   TM values: Max =    17  Min =   -26  Ave =    -4.55  Std =     7.33
   Min TM value:-26 at an offset of 121.850 usec -> 2018-05-19T02:14:08.473.926.600 (XO: 0.4740683)
   Max TM value:17 at an offset of 163.913 usec -> 2018-05-19T02:14:08.473.968.663 (XO: 0.4741104)
   Max dTM/dt [1/us]::2.76e+02 at an offset of 141.101 usec -> 2018-05-19T02:14:08.473.945.850 (XO: 0.4740875)
   Min dTM/dt [1/us]::-2.76e+02 at an offset of 161.638 usec -> 2018-05-19T02:14:08.473.966.388 (XO: 0.4741081)

 - Block number: 2 starting at 2018-05-19T02:14:32.366.711.750 (XO: 24.3668534, t=34.0°C)  Rank: 00008
   TM values: Max =    15  Min =   -24  Ave =    -4.47  Std =     6.87
   Min TM value:-24 at an offset of 112.013 usec -> 2018-05-19T02:14:32.366.823.762 (XO: 24.3669654)
   Max TM value:15 at an offset of 120.000 usec -> 2018-05-19T02:14:32.366.831.750 (XO: 24.3669734)
   Max dTM/dt [1/us]::2.25e+02 at an offset of 133.563 usec -> 2018-05-19T02:14:32.366.845.313 (XO: 24.3669870)
   Min dTM/dt [1/us]::-2.39e+02 at an offset of 106.763 usec -> 2018-05-19T02:14:32.366.818.512 (XO: 24.3669602)

 - Block number: 3 starting at 2018-05-19T02:14:50.738.733.750 (XO: 42.7388753, t=34.0°C)  Rank: 00005
   TM values: Max =    12  Min =   -22  Ave =    -4.51  Std =     5.68
   Min TM value:-22 at an offset of 105.400 usec -> 2018-05-19T02:14:50.738.839.150 (XO: 42.7389807)
   Max TM value:12 at an offset of 108.263 usec -> 2018-05-19T02:14:50.738.842.012 (XO: 42.7389836)
   Max dTM/dt [1/us]::2.00e+02 at an offset of 105.725 usec -> 2018-05-19T02:14:50.738.839.475 (XO: 42.7389810)
   Min dTM/dt [1/us]::-1.93e+02 at an offset of 119.263 usec -> 2018-05-19T02:14:50.738.853.012 (XO: 42.7389946)
