-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180518_073413_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.957093 to 30.348833
-each separately from 10s to 30s and from 348544us to 957382us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.703416 in 1 packets - #/No: 18/18
      4.687457 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-18T07:34:26.957.093.000 to 2018-05-18T07:34:46.348.833.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.703416 in 1 packets - #/No: 18/18
      4.687457 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4347707 to 31.4346932
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999419 in 1 packets - #/No: 18/18
      4.9999806 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180518_073413_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.3918612 to 12.3921502
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7033590  -- OK for SL --
   MIU time from 2018-05-18T07:34:26.957.093.000 to 2018-05-18T07:34:26.957.382.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.703416

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.0955092 to 27.0957982
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.6874390  -- OK for SL --
   MIU time from 2018-05-18T07:34:41.660.798.000 to 2018-05-18T07:34:41.661.087.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.687457

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7832372 to 31.7835262
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-18T07:34:46.348.544.000 to 2018-05-18T07:34:46.348.833.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-18T07:34:26.956.952.750
   Max SL corrected packet time: 2018-05-18T07:34:46.348.692.750
   Min SL corrected XO rel time: 10.9570942
   Max SL corrected XO rel time: 30.3488344
   TM values from -42 to 34 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-18T07:34:26.956.952.750 (XO: 10.9570942, t=34.0°C)  Rank: 00010
   TM values: Max =    34  Min =   -42  Ave =    -4.50  Std =    11.17
   Min TM value:-42 at an offset of 156.813 usec -> 2018-05-18T07:34:26.957.109.563 (XO: 10.9572510)
   Max TM value:34 at an offset of 145.613 usec -> 2018-05-18T07:34:26.957.098.363 (XO: 10.9572398)
   Max dTM/dt [1/us]::1.89e+02 at an offset of 93.350 usec -> 2018-05-18T07:34:26.957.046.100 (XO: 10.9571876)
   Min dTM/dt [1/us]::-2.11e+02 at an offset of 305.989 usec -> 2018-05-18T07:34:26.957.258.738 (XO: 10.9574002)

 - Block number: 2 starting at 2018-05-18T07:34:41.660.657.750 (XO: 25.6607992, t=34.0°C)  Rank: 00013
   TM values: Max =    25  Min =   -36  Ave =    -4.58  Std =    11.45
   Min TM value:-36 at an offset of 138.563 usec -> 2018-05-18T07:34:41.660.796.313 (XO: 25.6609377)
   Max TM value:25 at an offset of 136.288 usec -> 2018-05-18T07:34:41.660.794.038 (XO: 25.6609355)
   Max dTM/dt [1/us]::9.30e+01 at an offset of 149.576 usec -> 2018-05-18T07:34:41.660.807.325 (XO: 25.6609488)
   Min dTM/dt [1/us]::-2.43e+02 at an offset of 305.989 usec -> 2018-05-18T07:34:41.660.963.738 (XO: 25.6611052)

 - Block number: 3 starting at 2018-05-18T07:34:46.348.403.750 (XO: 30.3485454, t=34.0°C)  Rank: 00011
   TM values: Max =    29  Min =   -37  Ave =    -4.51  Std =    10.93
   Min TM value:-37 at an offset of 133.451 usec -> 2018-05-18T07:34:46.348.537.200 (XO: 30.3486788)
   Max TM value:29 at an offset of 153.913 usec -> 2018-05-18T07:34:46.348.557.663 (XO: 30.3486993)
   Max dTM/dt [1/us]::1.32e+02 at an offset of 155.301 usec -> 2018-05-18T07:34:46.348.559.050 (XO: 30.3487007)
   Min dTM/dt [1/us]::-1.39e+02 at an offset of 169.651 usec -> 2018-05-18T07:34:46.348.573.400 (XO: 30.3487150)
