-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180518_070357_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.750516 to 41.728383
-each separately from 1s to 41s and from 308312us to 750805us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.557507 in 1 packets - #/No: 18/18
      26.419493 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-18T07:04:01.750.516.000 to 2018-05-18T07:04:41.728.383.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.557507 in 1 packets - #/No: 18/18
      26.419493 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4294603 to 42.4293056
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999458 in 1 packets - #/No: 18/18
      25.9998995 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180518_070357_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.1799746 to 3.1802636
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.5574550  -- OK for SL --
   MIU time from 2018-05-18T07:04:01.750.516.000 to 2018-05-18T07:04:01.750.805.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.557507

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7377186 to 16.7380076
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.4193910  -- OK for SL --
   MIU time from 2018-05-18T07:04:15.308.312.000 to 2018-05-18T07:04:15.308.601.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.419493

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.1573986 to 43.1576876
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-18T07:04:41.728.094.000 to 2018-05-18T07:04:41.728.383.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-18T07:04:01.750.375.750
   Max SL corrected packet time: 2018-05-18T07:04:41.728.242.750
   Min SL corrected XO rel time: 1.7505172
   Max SL corrected XO rel time: 41.7283848
   TM values from -32 to 25 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-18T07:04:01.750.375.750 (XO: 1.7505172, t=34.0°C)  Rank: 00006
   TM values: Max =    25  Min =   -32  Ave =    -4.57  Std =     7.63
   Min TM value:-32 at an offset of 88.750 usec -> 2018-05-18T07:04:01.750.464.500 (XO: 1.7506060)
   Max TM value:25 at an offset of 88.125 usec -> 2018-05-18T07:04:01.750.463.875 (XO: 1.7506054)
   Max dTM/dt [1/us]::1.56e+02 at an offset of 110.800 usec -> 2018-05-18T07:04:01.750.486.550 (XO: 1.7506280)
   Min dTM/dt [1/us]::-1.46e+02 at an offset of 150.901 usec -> 2018-05-18T07:04:01.750.526.650 (XO: 1.7506681)

 - Block number: 2 starting at 2018-05-18T07:04:15.308.171.750 (XO: 15.3083137, t=34.0°C)  Rank: 00004
   TM values: Max =    16  Min =   -26  Ave =    -4.45  Std =     5.71
   Min TM value:-26 at an offset of 144.826 usec -> 2018-05-18T07:04:15.308.316.575 (XO: 15.3084585)
   Max TM value:16 at an offset of 81.275 usec -> 2018-05-18T07:04:15.308.253.025 (XO: 15.3083950)
   Max dTM/dt [1/us]::1.31e+02 at an offset of 305.989 usec -> 2018-05-18T07:04:15.308.477.738 (XO: 15.3086197)
   Min dTM/dt [1/us]::-1.11e+02 at an offset of 173.076 usec -> 2018-05-18T07:04:15.308.344.825 (XO: 15.3084868)

 - Block number: 3 starting at 2018-05-18T07:04:41.727.953.750 (XO: 41.7280958, t=34.0°C)  Rank: 00006
   TM values: Max =    23  Min =   -32  Ave =    -4.49  Std =     8.44
   Min TM value:-32 at an offset of 151.888 usec -> 2018-05-18T07:04:41.728.105.638 (XO: 41.7282477)
   Max TM value:23 at an offset of 188.051 usec -> 2018-05-18T07:04:41.728.141.800 (XO: 41.7282839)
   Max dTM/dt [1/us]::1.85e+02 at an offset of 170.988 usec -> 2018-05-18T07:04:41.728.124.738 (XO: 41.7282668)
   Min dTM/dt [1/us]::-1.78e+02 at an offset of 190.001 usec -> 2018-05-18T07:04:41.728.143.750 (XO: 41.7282858)
