-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180517_035100_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.175007 to 34.668467
-each separately from 0s to 34s and from 175007us to 928438us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.752853 in 1 packets - #/No: 18/18
      7.739740 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-17T03:51:03.175.007.000 to 2018-05-17T03:51:37.668.467.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      26.752853 in 1 packets - #/No: 18/18
      7.739740 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4384994 to 35.4383695
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      25.9999007 in 1 packets - #/No: 18/18
      7.9999694 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180517_035100_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.6135070 to 1.6137960
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.7527510  -- OK for SL --
   MIU time from 2018-05-17T03:51:03.175.007.000 to 2018-05-17T03:51:03.175.296.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.752853

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.3665470 to 28.3668360
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7397110  -- OK for SL --
   MIU time from 2018-05-17T03:51:29.928.149.000 to 2018-05-17T03:51:29.928.438.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.739740

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.1065470 to 36.1068360
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-17T03:51:37.668.178.000 to 2018-05-17T03:51:37.668.467.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-17T03:51:03.174.866.750
   Max SL corrected packet time: 2018-05-17T03:51:37.668.326.750
   Min SL corrected XO rel time: 0.1750083
   Max SL corrected XO rel time: 34.6684690
   TM values from -27 to 17 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-17T03:51:03.174.866.750 (XO: 0.1750083, t=34.0°C)  Rank: 00006
   TM values: Max =    17  Min =   -27  Ave =    -4.59  Std =     7.80
   Min TM value:-27 at an offset of 136.363 usec -> 2018-05-17T03:51:03.175.003.113 (XO: 0.1751446)
   Max TM value:17 at an offset of 127.763 usec -> 2018-05-17T03:51:03.174.994.512 (XO: 0.1751360)
   Max dTM/dt [1/us]::3.92e+02 at an offset of 127.550 usec -> 2018-05-17T03:51:03.174.994.300 (XO: 0.1751358)
   Min dTM/dt [1/us]::-4.02e+02 at an offset of 127.800 usec -> 2018-05-17T03:51:03.174.994.550 (XO: 0.1751361)

 - Block number: 2 starting at 2018-05-17T03:51:29.928.008.750 (XO: 26.9281505, t=34.0°C)  Rank: 00005
   TM values: Max =    12  Min =   -22  Ave =    -4.51  Std =     5.42
   Min TM value:-22 at an offset of 170.663 usec -> 2018-05-17T03:51:29.928.179.413 (XO: 26.9283212)
   Max TM value:12 at an offset of 144.863 usec -> 2018-05-17T03:51:29.928.153.613 (XO: 26.9282954)
   Max dTM/dt [1/us]::3.09e+02 at an offset of 144.826 usec -> 2018-05-17T03:51:29.928.153.575 (XO: 26.9282953)
   Min dTM/dt [1/us]::-3.03e+02 at an offset of 131.638 usec -> 2018-05-17T03:51:29.928.140.388 (XO: 26.9282822)

 - Block number: 3 starting at 2018-05-17T03:51:37.668.037.750 (XO: 34.6681800, t=34.0°C)  Rank: 00005
   TM values: Max =    14  Min =   -23  Ave =    -4.57  Std =     5.34
   Min TM value:-23 at an offset of 110.263 usec -> 2018-05-17T03:51:37.668.148.012 (XO: 34.6682903)
   Max TM value:14 at an offset of 136.376 usec -> 2018-05-17T03:51:37.668.174.125 (XO: 34.6683164)
   Max dTM/dt [1/us]::3.34e+02 at an offset of 110.300 usec -> 2018-05-17T03:51:37.668.148.050 (XO: 34.6682903)
   Min dTM/dt [1/us]::-3.32e+02 at an offset of 127.575 usec -> 2018-05-17T03:51:37.668.165.325 (XO: 34.6683076)
