-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180515_190205_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.510808 to 38.057610
-each separately from 12s to 38s and from 57321us to 648567us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.137181 in 1 packets - #/No: 18/18
      13.408754 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-15T19:02:20.510.808.000 to 2018-05-15T19:02:46.057.610.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.137181 in 1 packets - #/No: 18/18
      13.408754 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4370169 to 39.4369190
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999548 in 1 packets - #/No: 18/18
      13.9999473 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180515_190205_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      24 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9478248 to 13.9481138
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1371350  -- OK for SL --
   MIU time from 2018-05-15T19:02:20.510.808.000 to 2018-05-15T19:02:20.511.097.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.137181

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.0852488 to 26.0855378
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4087030  -- OK for SL --
   MIU time from 2018-05-15T19:02:32.648.278.000 to 2018-05-15T19:02:32.648.567.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.408754

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.4942408 to 39.4945298
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-15T19:02:46.057.321.000 to 2018-05-15T19:02:46.057.610.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-15T19:02:20.510.667.750
   Max SL corrected packet time: 2018-05-15T19:02:46.057.469.750
   Min SL corrected XO rel time: 12.5108098
   Max SL corrected XO rel time: 38.0576110
   TM values from -92 to 99 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-15T19:02:20.510.667.750 (XO: 12.5108098, t=34.0°C)  Rank: 00024
   TM values: Max =    99  Min =   -92  Ave =    -4.52  Std =    14.60
   Min TM value:-92 at an offset of 111.363 usec -> 2018-05-15T19:02:20.510.779.112 (XO: 12.5109212)
   Max TM value:99 at an offset of 102.250 usec -> 2018-05-15T19:02:20.510.770.000 (XO: 12.5109121)
   Max dTM/dt [1/us]::9.03e+02 at an offset of 116.463 usec -> 2018-05-15T19:02:20.510.784.212 (XO: 12.5109263)
   Min dTM/dt [1/us]::-9.82e+02 at an offset of 106.088 usec -> 2018-05-15T19:02:20.510.773.837 (XO: 12.5109159)

 - Block number: 2 starting at 2018-05-15T19:02:32.648.137.750 (XO: 24.6482796, t=34.0°C)  Rank: 00013
   TM values: Max =    26  Min =   -35  Ave =    -4.51  Std =     7.96
   Min TM value:-35 at an offset of 116.888 usec -> 2018-05-15T19:02:32.648.254.637 (XO: 24.6483965)
   Max TM value:26 at an offset of 116.000 usec -> 2018-05-15T19:02:32.648.253.750 (XO: 24.6483956)
   Max dTM/dt [1/us]::3.61e+02 at an offset of 113.750 usec -> 2018-05-15T19:02:32.648.251.500 (XO: 24.6483933)
   Min dTM/dt [1/us]::-3.63e+02 at an offset of 115.913 usec -> 2018-05-15T19:02:32.648.253.662 (XO: 24.6483955)

 - Block number: 3 starting at 2018-05-15T19:02:46.057.180.750 (XO: 38.0573220, t=34.0°C)  Rank: 00010
   TM values: Max =    27  Min =   -38  Ave =    -4.44  Std =     8.96
   Min TM value:-38 at an offset of 124.463 usec -> 2018-05-15T19:02:46.057.305.212 (XO: 38.0574465)
   Max TM value:27 at an offset of 168.088 usec -> 2018-05-15T19:02:46.057.348.838 (XO: 38.0574901)
   Max dTM/dt [1/us]::3.64e+02 at an offset of 125.013 usec -> 2018-05-15T19:02:46.057.305.762 (XO: 38.0574470)
   Min dTM/dt [1/us]::-3.55e+02 at an offset of 123.063 usec -> 2018-05-15T19:02:46.057.303.812 (XO: 38.0574451)
