-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180515_105749_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.560907 to 35.532318
-each separately from 13s to 35s and from 129455us to 561196us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.568259 in 1 packets - #/No: 18/18
      12.402285 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-15T10:58:05.560.907.000 to 2018-05-15T10:58:27.532.318.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.568259 in 1 packets - #/No: 18/18
      12.402285 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4330573 to 36.4329745
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999623 in 1 packets - #/No: 18/18
      11.9999549 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180515_105749_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.9939633 to 14.9942523
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5682230  -- OK for SL --
   MIU time from 2018-05-15T10:58:05.560.907.000 to 2018-05-15T10:58:05.561.196.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.568259

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.5624753 to 24.5627643
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4022390  -- OK for SL --
   MIU time from 2018-05-15T10:58:15.129.455.000 to 2018-05-15T10:58:15.129.744.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.402285

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.9650033 to 36.9652923
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-15T10:58:27.532.029.000 to 2018-05-15T10:58:27.532.318.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-15T10:58:05.560.766.750
   Max SL corrected packet time: 2018-05-15T10:58:27.532.177.750
   Min SL corrected XO rel time: 13.5609081
   Max SL corrected XO rel time: 35.5323199
   TM values from -18 to 14 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-15T10:58:05.560.766.750 (XO: 13.5609081, t=34.0°C)  Rank: 00005
   TM values: Max =     4  Min =   -13  Ave =    -4.54  Std =     2.44
   Min TM value:-13 at an offset of 37.488 usec -> 2018-05-15T10:58:05.560.804.237 (XO: 13.5609456)
   Max TM value:4 at an offset of 73.363 usec -> 2018-05-15T10:58:05.560.840.112 (XO: 13.5609814)
   Max dTM/dt [1/us]::6.00e+01 at an offset of 266.201 usec -> 2018-05-15T10:58:05.561.032.950 (XO: 13.5611743)
   Min dTM/dt [1/us]::-6.20e+01 at an offset of 253.088 usec -> 2018-05-15T10:58:05.561.019.838 (XO: 13.5611612)

 - Block number: 2 starting at 2018-05-15T10:58:15.129.314.750 (XO: 23.1294562, t=34.0°C)  Rank: 00002
   TM values: Max =     0  Min =    -9  Ave =    -4.60  Std =     1.19
   Min TM value:-9 at an offset of 11.350 usec -> 2018-05-15T10:58:15.129.326.100 (XO: 23.1294675)
   Max TM value:0 at an offset of 68.763 usec -> 2018-05-15T10:58:15.129.383.512 (XO: 23.1295249)
   Max dTM/dt [1/us]::4.30e+01 at an offset of 194.263 usec -> 2018-05-15T10:58:15.129.509.013 (XO: 23.1296504)
   Min dTM/dt [1/us]::-3.90e+01 at an offset of 141.913 usec -> 2018-05-15T10:58:15.129.456.663 (XO: 23.1295981)

 - Block number: 3 starting at 2018-05-15T10:58:27.531.888.750 (XO: 35.5320309, t=34.0°C)  Rank: 00004
   TM values: Max =    14  Min =   -18  Ave =    -4.48  Std =     2.53
   Min TM value:-18 at an offset of 115.613 usec -> 2018-05-15T10:58:27.532.004.362 (XO: 35.5321465)
   Max TM value:14 at an offset of 113.188 usec -> 2018-05-15T10:58:27.532.001.937 (XO: 35.5321441)
   Max dTM/dt [1/us]::5.10e+01 at an offset of 67.225 usec -> 2018-05-15T10:58:27.531.955.975 (XO: 35.5320981)
   Min dTM/dt [1/us]::-5.50e+01 at an offset of 1.225 usec -> 2018-05-15T10:58:27.531.889.975 (XO: 35.5320321)
