-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180512_005031_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.473896 to 34.983199
-each separately from 12s to 34s and from 10994us to 983199us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.536809 in 1 packets - #/No: 18/18
      19.971627 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-12T00:50:46.473.896.000 to 2018-05-12T00:51:08.983.199.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.536809 in 1 packets - #/No: 18/18
      19.971627 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4298953 to 35.4298112
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.9999885 in 1 packets - #/No: 18/18
      18.9999274 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180512_005031_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9037909 to 13.9040799
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.5367990  -- OK for SL --
   MIU time from 2018-05-12T00:50:46.473.896.000 to 2018-05-12T00:50:46.474.185.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.536809

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4408789 to 16.4411679
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.9715510  -- OK for SL --
   MIU time from 2018-05-12T00:50:49.010.994.000 to 2018-05-12T00:50:49.011.283.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.971627

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.4127189 to 36.4130079
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-12T00:51:08.982.910.000 to 2018-05-12T00:51:08.983.199.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-12T00:50:46.473.755.750
   Max SL corrected packet time: 2018-05-12T00:51:08.983.058.750
   Min SL corrected XO rel time: 12.4738974
   Max SL corrected XO rel time: 34.9832004
   TM values from -13 to 6 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-12T00:50:46.473.755.750 (XO: 12.4738974, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -13  Ave =    -4.48  Std =     2.52
   Min TM value:-13 at an offset of 40.875 usec -> 2018-05-12T00:50:46.473.796.625 (XO: 12.4739383)
   Max TM value:6 at an offset of 127.650 usec -> 2018-05-12T00:50:46.473.883.400 (XO: 12.4740251)
   Max dTM/dt [1/us]::1.35e+02 at an offset of 127.600 usec -> 2018-05-12T00:50:46.473.883.350 (XO: 12.4740250)
   Min dTM/dt [1/us]::-1.31e+02 at an offset of 117.725 usec -> 2018-05-12T00:50:46.473.873.475 (XO: 12.4740152)

 - Block number: 2 starting at 2018-05-12T00:50:49.010.853.750 (XO: 15.0109951, t=34.0°C)  Rank: 00001
   TM values: Max =     3  Min =   -12  Ave =    -4.56  Std =     2.08
   Min TM value:-12 at an offset of 4.775 usec -> 2018-05-12T00:50:49.010.858.525 (XO: 15.0109999)
   Max TM value:3 at an offset of 170.463 usec -> 2018-05-12T00:50:49.011.024.213 (XO: 15.0111656)
   Max dTM/dt [1/us]::1.11e+02 at an offset of 182.038 usec -> 2018-05-12T00:50:49.011.035.788 (XO: 15.0111772)
   Min dTM/dt [1/us]::-1.14e+02 at an offset of 179.788 usec -> 2018-05-12T00:50:49.011.033.538 (XO: 15.0111749)

 - Block number: 3 starting at 2018-05-12T00:51:08.982.769.750 (XO: 34.9829114, t=34.0°C)  Rank: 00002
   TM values: Max =     3  Min =   -12  Ave =    -4.49  Std =     2.16
   Min TM value:-12 at an offset of 167.851 usec -> 2018-05-12T00:51:08.982.937.600 (XO: 34.9830793)
   Max TM value:3 at an offset of 15.088 usec -> 2018-05-12T00:51:08.982.784.837 (XO: 34.9829265)
   Max dTM/dt [1/us]::1.13e+02 at an offset of 160.938 usec -> 2018-05-12T00:51:08.982.930.688 (XO: 34.9830724)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 158.676 usec -> 2018-05-12T00:51:08.982.928.425 (XO: 34.9830701)
