-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180507_034446_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.891471 to 34.990682
-each separately from 6s to 34s and from 888070us to 990682us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.996310 in 1 packets - #/No: 18/18
      5.102034 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-07T03:44:55.891.471.000 to 2018-05-07T03:45:23.990.682.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.996310 in 1 packets - #/No: 18/18
      5.102034 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4400444 to 35.4399389
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.9999134 in 1 packets - #/No: 18/18
      4.9999811 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180507_034446_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.3315141 to 8.3318031
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.9962230  -- OK for SL --
   MIU time from 2018-05-07T03:44:55.891.471.000 to 2018-05-07T03:44:55.891.760.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.996310

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.3280261 to 31.3283151
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.1020150  -- OK for SL --
   MIU time from 2018-05-07T03:45:18.888.070.000 to 2018-05-07T03:45:18.888.359.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.102034

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.4303301 to 36.4306191
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-07T03:45:23.990.393.000 to 2018-05-07T03:45:23.990.682.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-07T03:44:55.891.330.750
   Max SL corrected packet time: 2018-05-07T03:45:23.990.541.750
   Min SL corrected XO rel time: 6.8914730
   Max SL corrected XO rel time: 34.9906840
   TM values from -79 to 47 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-07T03:44:55.891.330.750 (XO: 6.8914730, t=34.0°C)  Rank: 00008
   TM values: Max =    41  Min =   -74  Ave =    -4.33  Std =     4.21
   Min TM value:-74 at an offset of 99.225 usec -> 2018-05-07T03:44:55.891.429.975 (XO: 6.8915722)
   Max TM value:41 at an offset of 99.550 usec -> 2018-05-07T03:44:55.891.430.300 (XO: 6.8915725)
   Max dTM/dt [1/us]::6.44e+02 at an offset of 93.625 usec -> 2018-05-07T03:44:55.891.424.375 (XO: 6.8915666)
   Min dTM/dt [1/us]::-5.32e+02 at an offset of 88.963 usec -> 2018-05-07T03:44:55.891.419.712 (XO: 6.8915620)

 - Block number: 2 starting at 2018-05-07T03:45:18.887.929.750 (XO: 29.8880717, t=34.0°C)  Rank: 00009
   TM values: Max =    29  Min =   -59  Ave =    -4.47  Std =     5.08
   Min TM value:-59 at an offset of 68.300 usec -> 2018-05-07T03:45:18.887.998.050 (XO: 29.8881400)
   Max TM value:29 at an offset of 70.675 usec -> 2018-05-07T03:45:18.888.000.425 (XO: 29.8881423)
   Max dTM/dt [1/us]::7.17e+02 at an offset of 70.625 usec -> 2018-05-07T03:45:18.888.000.375 (XO: 29.8881423)
   Min dTM/dt [1/us]::-4.21e+02 at an offset of 178.101 usec -> 2018-05-07T03:45:18.888.107.850 (XO: 29.8882498)

 - Block number: 3 starting at 2018-05-07T03:45:23.990.252.750 (XO: 34.9903950, t=34.0°C)  Rank: 00008
   TM values: Max =    47  Min =   -79  Ave =    -4.40  Std =     6.27
   Min TM value:-79 at an offset of 177.613 usec -> 2018-05-07T03:45:23.990.430.363 (XO: 34.9905726)
   Max TM value:47 at an offset of 289.339 usec -> 2018-05-07T03:45:23.990.542.088 (XO: 34.9906843)
   Max dTM/dt [1/us]::9.75e+02 at an offset of 12.750 usec -> 2018-05-07T03:45:23.990.265.500 (XO: 34.9904077)
   Min dTM/dt [1/us]::-7.54e+02 at an offset of 112.413 usec -> 2018-05-07T03:45:23.990.365.162 (XO: 34.9905074)
