-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180505_225919_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.528132 to 43.645605
-each separately from 1s to 43s and from 66582us to 645605us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.538161 in 1 packets - #/No: 18/18
      24.578445 in 1 packets - #/No: 36/18
MIU packet time from 2018-05-05T22:59:23.528.132.000 to 2018-05-05T23:00:05.645.605.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.538161 in 1 packets - #/No: 18/18
      24.578445 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4370115 to 44.4368527
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999319 in 1 packets - #/No: 18/18
      23.9999093 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180505_225919_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      45 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.9651425 to 2.9654315
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5380950  -- OK for SL --
   MIU time from 2018-05-05T22:59:23.528.132.000 to 2018-05-05T22:59:23.528.421.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.538161

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      32 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.5035265 to 20.5038155
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.5783510  -- OK for SL --
   MIU time from 2018-05-05T22:59:41.066.582.000 to 2018-05-05T22:59:41.066.871.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.578445

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      18 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.0821665 to 45.0824555
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-05-05T23:00:05.645.316.000 to 2018-05-05T23:00:05.645.605.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-05-05T22:59:23.527.991.750
   Max SL corrected packet time: 2018-05-05T23:00:05.645.464.750
   Min SL corrected XO rel time: 1.5281330
   Max SL corrected XO rel time: 43.6456052
   TM values from -70 to 60 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-05-05T22:59:23.527.991.750 (XO: 1.5281330, t=34.0°C)  Rank: 00045
   TM values: Max =    60  Min =   -70  Ave =    -4.56  Std =    23.15
   Min TM value:-70 at an offset of 145.151 usec -> 2018-05-05T22:59:23.528.136.900 (XO: 1.5282782)
   Max TM value:60 at an offset of 144.263 usec -> 2018-05-05T22:59:23.528.136.013 (XO: 1.5282773)
   Max dTM/dt [1/us]::1.15e+03 at an offset of 132.563 usec -> 2018-05-05T22:59:23.528.124.313 (XO: 1.5282656)
   Min dTM/dt [1/us]::-1.13e+03 at an offset of 145.113 usec -> 2018-05-05T22:59:23.528.136.863 (XO: 1.5282781)

 - Block number: 2 starting at 2018-05-05T22:59:41.066.441.750 (XO: 19.0665834, t=34.0°C)  Rank: 00032
   TM values: Max =    55  Min =   -66  Ave =    -4.53  Std =    24.01
   Min TM value:-66 at an offset of 138.001 usec -> 2018-05-05T22:59:41.066.579.750 (XO: 19.0667214)
   Max TM value:55 at an offset of 144.726 usec -> 2018-05-05T22:59:41.066.586.475 (XO: 19.0667281)
   Max dTM/dt [1/us]::1.07e+03 at an offset of 138.038 usec -> 2018-05-05T22:59:41.066.579.788 (XO: 19.0667214)
   Min dTM/dt [1/us]::-1.09e+03 at an offset of 152.538 usec -> 2018-05-05T22:59:41.066.594.288 (XO: 19.0667359)

 - Block number: 3 starting at 2018-05-05T23:00:05.645.175.750 (XO: 43.6453162, t=34.0°C)  Rank: 00018
   TM values: Max =    38  Min =   -47  Ave =    -4.59  Std =    13.06
   Min TM value:-47 at an offset of 127.650 usec -> 2018-05-05T23:00:05.645.303.400 (XO: 43.6454438)
   Max TM value:38 at an offset of 140.201 usec -> 2018-05-05T23:00:05.645.315.950 (XO: 43.6454564)
   Max dTM/dt [1/us]::7.01e+02 at an offset of 125.888 usec -> 2018-05-05T23:00:05.645.301.637 (XO: 43.6454421)
   Min dTM/dt [1/us]::-6.93e+02 at an offset of 148.976 usec -> 2018-05-05T23:00:05.645.324.725 (XO: 43.6454652)
