-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180430_162118_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.711273 to 32.886264
-each separately from 11s to 32s and from 602897us to 886264us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.891335 in 1 packets - #/No: 18/18
      10.282789 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-30T16:21:32.711.273.000 to 2018-04-30T16:21:53.886.264.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.891335 in 1 packets - #/No: 18/18
      10.282789 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.4412938 to 33.4412155
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.9999590 in 1 packets - #/No: 18/18
      9.9999627 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180430_162118_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.1525653 to 13.1528543
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8912950  -- OK for SL --
   MIU time from 2018-04-30T16:21:32.711.273.000 to 2018-04-30T16:21:32.711.562.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.891335

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.0441493 to 24.0444383
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.2827510  -- OK for SL --
   MIU time from 2018-04-30T16:21:43.602.897.000 to 2018-04-30T16:21:43.603.186.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.282789

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.3271893 to 34.3274783
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-30T16:21:53.885.975.000 to 2018-04-30T16:21:53.886.264.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-30T16:21:32.711.132.750
   Max SL corrected packet time: 2018-04-30T16:21:53.886.123.750
   Min SL corrected XO rel time: 11.7112741
   Max SL corrected XO rel time: 32.8862661
   TM values from -22 to 13 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-30T16:21:32.711.132.750 (XO: 11.7112741, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -19  Ave =    -4.47  Std =     4.26
   Min TM value:-19 at an offset of 151.913 usec -> 2018-04-30T16:21:32.711.284.663 (XO: 11.7114260)
   Max TM value:10 at an offset of 67.613 usec -> 2018-04-30T16:21:32.711.200.362 (XO: 11.7113417)
   Max dTM/dt [1/us]::1.53e+02 at an offset of 177.938 usec -> 2018-04-30T16:21:32.711.310.688 (XO: 11.7114521)
   Min dTM/dt [1/us]::-1.55e+02 at an offset of 12.713 usec -> 2018-04-30T16:21:32.711.145.462 (XO: 11.7112868)

 - Block number: 2 starting at 2018-04-30T16:21:43.602.756.750 (XO: 22.6028987, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -20  Ave =    -4.65  Std =     4.29
   Min TM value:-20 at an offset of 125.125 usec -> 2018-04-30T16:21:43.602.881.875 (XO: 22.6030239)
   Max TM value:12 at an offset of 118.313 usec -> 2018-04-30T16:21:43.602.875.062 (XO: 22.6030170)
   Max dTM/dt [1/us]::1.66e+02 at an offset of 118.288 usec -> 2018-04-30T16:21:43.602.875.037 (XO: 22.6030170)
   Min dTM/dt [1/us]::-1.50e+02 at an offset of 63.588 usec -> 2018-04-30T16:21:43.602.820.337 (XO: 22.6029623)

 - Block number: 3 starting at 2018-04-30T16:21:53.885.834.750 (XO: 32.8859771, t=34.0°C)  Rank: 00003
   TM values: Max =    13  Min =   -22  Ave =    -4.57  Std =     4.38
   Min TM value:-22 at an offset of 107.413 usec -> 2018-04-30T16:21:53.885.942.162 (XO: 32.8860845)
   Max TM value:13 at an offset of 106.863 usec -> 2018-04-30T16:21:53.885.941.612 (XO: 32.8860839)
   Max dTM/dt [1/us]::1.21e+02 at an offset of 107.338 usec -> 2018-04-30T16:21:53.885.942.087 (XO: 32.8860844)
   Min dTM/dt [1/us]::-1.20e+02 at an offset of 118.275 usec -> 2018-04-30T16:21:53.885.953.025 (XO: 32.8860954)
