-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180427_001619_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 9.899066 to 42.772563
-each separately from 9s to 42s and from 737219us to 899355us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.837864 in 1 packets - #/No: 18/18
      26.034766 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-27T00:16:31.899.066.000 to 2018-04-27T00:17:04.772.563.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.837864 in 1 packets - #/No: 18/18
      26.034766 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4323423 to 43.4322213
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999743 in 1 packets - #/No: 18/18
      25.9999047 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180427_001619_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.3314062 to 11.3316952
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.8378390  -- OK for SL --
   MIU time from 2018-04-27T00:16:31.899.066.000 to 2018-04-27T00:16:31.899.355.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.837864

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.1695342 to 18.1698232
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.0346710  -- OK for SL --
   MIU time from 2018-04-27T00:16:38.737.219.000 to 2018-04-27T00:16:38.737.508.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.034766

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.2044942 to 44.2047832
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-27T00:17:04.772.274.000 to 2018-04-27T00:17:04.772.563.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-27T00:16:31.898.925.750
   Max SL corrected packet time: 2018-04-27T00:17:04.772.422.750
   Min SL corrected XO rel time: 9.8990672
   Max SL corrected XO rel time: 42.7725648
   TM values from -16 to 8 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-27T00:16:31.898.925.750 (XO: 9.8990672, t=34.0°C)  Rank: 00002
   TM values: Max =     4  Min =   -13  Ave =    -4.49  Std =     2.52
   Min TM value:-13 at an offset of 82.225 usec -> 2018-04-27T00:16:31.899.007.975 (XO: 9.8991495)
   Max TM value:4 at an offset of 94.413 usec -> 2018-04-27T00:16:31.899.020.162 (XO: 9.8991616)
   Max dTM/dt [1/us]::1.03e+02 at an offset of 82.538 usec -> 2018-04-27T00:16:31.899.008.287 (XO: 9.8991498)
   Min dTM/dt [1/us]::-1.01e+02 at an offset of 151.451 usec -> 2018-04-27T00:16:31.899.077.200 (XO: 9.8992187)

 - Block number: 2 starting at 2018-04-27T00:16:38.737.078.750 (XO: 16.7372203, t=34.0°C)  Rank: 00002
   TM values: Max =     5  Min =   -13  Ave =    -4.50  Std =     2.35
   Min TM value:-13 at an offset of 105.438 usec -> 2018-04-27T00:16:38.737.184.187 (XO: 16.7373257)
   Max TM value:5 at an offset of 2.388 usec -> 2018-04-27T00:16:38.737.081.137 (XO: 16.7372226)
   Max dTM/dt [1/us]::9.30e+01 at an offset of 190.651 usec -> 2018-04-27T00:16:38.737.269.400 (XO: 16.7374109)
   Min dTM/dt [1/us]::-1.00e+02 at an offset of 200.651 usec -> 2018-04-27T00:16:38.737.279.400 (XO: 16.7374209)

 - Block number: 3 starting at 2018-04-27T00:17:04.772.133.750 (XO: 42.7722758, t=34.0°C)  Rank: 00002
   TM values: Max =     8  Min =   -16  Ave =    -4.52  Std =     4.28
   Min TM value:-16 at an offset of 33.438 usec -> 2018-04-27T00:17:04.772.167.187 (XO: 42.7723092)
   Max TM value:8 at an offset of 91.600 usec -> 2018-04-27T00:17:04.772.225.350 (XO: 42.7723674)
   Max dTM/dt [1/us]::1.56e+02 at an offset of 35.788 usec -> 2018-04-27T00:17:04.772.169.537 (XO: 42.7723115)
   Min dTM/dt [1/us]::-1.64e+02 at an offset of 91.650 usec -> 2018-04-27T00:17:04.772.225.400 (XO: 42.7723674)
