-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180424_130354_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.000093 to 30.022289
-each separately from 0s to 30s and from 93us to 22289us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.007917 in 1 packets - #/No: 18/18
      14.013412 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-24T13:03:46.000.093.000 to 2018-04-24T13:04:16.022.289.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.007917 in 1 packets - #/No: 18/18
      14.013412 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4311340 to 31.4310191
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999387 in 1 packets - #/No: 18/18
      13.9999464 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      BE -> 32.4°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180424_130354_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4312285 to 1.4315175
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.0078550  -- OK for SL --
   MIU time from 2018-04-24T13:03:46.000.093.000 to 2018-04-24T13:03:46.000.382.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.007917

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.4393725 to 17.4396615
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0133590  -- OK for SL --
   MIU time from 2018-04-24T13:04:02.008.299.000 to 2018-04-24T13:04:02.008.588.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.013412

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4530205 to 31.4533095
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-24T13:04:16.022.000.000 to 2018-04-24T13:04:16.022.289.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-24T13:03:45.999.952.750
   Max SL corrected packet time: 2018-04-24T13:04:16.022.148.750
   Min SL corrected XO rel time: 0.0000945
   Max SL corrected XO rel time: 30.0222905
   TM values from -13 to 4 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-24T13:03:45.999.952.750 (XO: 0.0000945, t=34.0°C)  Rank: 00001
   TM values: Max =     4  Min =   -13  Ave =    -4.45  Std =     2.43
   Min TM value:-13 at an offset of 225.851 usec -> 2018-04-24T13:03:46.000.178.600 (XO: 0.0003204)
   Max TM value:4 at an offset of 162.626 usec -> 2018-04-24T13:03:46.000.115.375 (XO: 0.0002571)
   Max dTM/dt [1/us]::6.40e+01 at an offset of 238.326 usec -> 2018-04-24T13:03:46.000.191.075 (XO: 0.0003328)
   Min dTM/dt [1/us]::-7.40e+01 at an offset of 225.801 usec -> 2018-04-24T13:03:46.000.178.550 (XO: 0.0003203)

 - Block number: 2 starting at 2018-04-24T13:04:02.008.158.750 (XO: 16.0082998, t=34.0°C)  Rank: 00003
   TM values: Max =     3  Min =   -12  Ave =    -4.47  Std =     2.29
   Min TM value:-12 at an offset of 120.963 usec -> 2018-04-24T13:04:02.008.279.712 (XO: 16.0084208)
   Max TM value:3 at an offset of 211.301 usec -> 2018-04-24T13:04:02.008.370.050 (XO: 16.0085111)
   Max dTM/dt [1/us]::6.00e+01 at an offset of 164.901 usec -> 2018-04-24T13:04:02.008.323.650 (XO: 16.0084647)
   Min dTM/dt [1/us]::-4.80e+01 at an offset of 161.413 usec -> 2018-04-24T13:04:02.008.320.163 (XO: 16.0084612)

 - Block number: 3 starting at 2018-04-24T13:04:16.021.859.750 (XO: 30.0220015, t=34.0°C)  Rank: 00001
   TM values: Max =     2  Min =   -10  Ave =    -4.52  Std =     1.35
   Min TM value:-10 at an offset of 24.888 usec -> 2018-04-24T13:04:16.021.884.637 (XO: 30.0220264)
   Max TM value:2 at an offset of 182.263 usec -> 2018-04-24T13:04:16.022.042.013 (XO: 30.0221837)
   Max dTM/dt [1/us]::6.10e+01 at an offset of 182.476 usec -> 2018-04-24T13:04:16.022.042.225 (XO: 30.0221840)
   Min dTM/dt [1/us]::-7.30e+01 at an offset of 182.313 usec -> 2018-04-24T13:04:16.022.042.063 (XO: 30.0221838)
