-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180423_165314_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.549448 to 37.761880
-each separately from 0s to 37s and from 324276us to 761880us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.774539 in 1 packets - #/No: 18/18
      17.437026 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-23T16:53:06.549.448.000 to 2018-04-23T16:53:43.761.880.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.774539 in 1 packets - #/No: 18/18
      17.437026 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4326040 to 38.4324610
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.9999227 in 1 packets - #/No: 18/18
      16.9999343 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180423_165314_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9820511 to 1.9823401
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.7744630  -- OK for SL --
   MIU time from 2018-04-23T16:53:06.549.448.000 to 2018-04-23T16:53:06.549.737.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.774539

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.7568031 to 21.7570921
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4369590  -- OK for SL --
   MIU time from 2018-04-23T16:53:26.324.276.000 to 2018-04-23T16:53:26.324.565.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.437026

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.1940511 to 39.1943401
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-23T16:53:43.761.591.000 to 2018-04-23T16:53:43.761.880.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-23T16:53:06.549.307.750
   Max SL corrected packet time: 2018-04-23T16:53:43.761.739.750
   Min SL corrected XO rel time: 0.5494492
   Max SL corrected XO rel time: 37.7618821
   TM values from -28 to 20 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-23T16:53:06.549.307.750 (XO: 0.5494492, t=34.0°C)  Rank: 00004
   TM values: Max =    20  Min =   -28  Ave =    -4.44  Std =     6.27
   Min TM value:-28 at an offset of 51.863 usec -> 2018-04-23T16:53:06.549.359.612 (XO: 0.5495011)
   Max TM value:20 at an offset of 124.313 usec -> 2018-04-23T16:53:06.549.432.062 (XO: 0.5495735)
   Max dTM/dt [1/us]::2.99e+02 at an offset of 124.275 usec -> 2018-04-23T16:53:06.549.432.025 (XO: 0.5495735)
   Min dTM/dt [1/us]::-2.78e+02 at an offset of 138.426 usec -> 2018-04-23T16:53:06.549.446.175 (XO: 0.5495876)

 - Block number: 2 starting at 2018-04-23T16:53:26.324.135.750 (XO: 20.3242777, t=34.0°C)  Rank: 00004
   TM values: Max =    17  Min =   -25  Ave =    -4.36  Std =     5.94
   Min TM value:-25 at an offset of 144.376 usec -> 2018-04-23T16:53:26.324.280.125 (XO: 20.3244220)
   Max TM value:17 at an offset of 110.325 usec -> 2018-04-23T16:53:26.324.246.075 (XO: 20.3243880)
   Max dTM/dt [1/us]::1.90e+02 at an offset of 127.713 usec -> 2018-04-23T16:53:26.324.263.462 (XO: 20.3244054)
   Min dTM/dt [1/us]::-1.97e+02 at an offset of 110.375 usec -> 2018-04-23T16:53:26.324.246.125 (XO: 20.3243880)

 - Block number: 3 starting at 2018-04-23T16:53:43.761.450.750 (XO: 37.7615931, t=34.0°C)  Rank: 00004
   TM values: Max =    18  Min =   -26  Ave =    -4.39  Std =     6.81
   Min TM value:-26 at an offset of 112.138 usec -> 2018-04-23T16:53:43.761.562.887 (XO: 37.7617052)
   Max TM value:18 at an offset of 115.875 usec -> 2018-04-23T16:53:43.761.566.625 (XO: 37.7617089)
   Max dTM/dt [1/us]::2.42e+02 at an offset of 64.900 usec -> 2018-04-23T16:53:43.761.515.650 (XO: 37.7616580)
   Min dTM/dt [1/us]::-1.89e+02 at an offset of 111.575 usec -> 2018-04-23T16:53:43.761.562.325 (XO: 37.7617046)
