-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180422_033328_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.426999 to 40.470336
-each separately from 8s to 40s and from 426999us to 833139us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.405562 in 1 packets - #/No: 18/18
      24.636908 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-22T03:33:28.426.999.000 to 2018-04-22T03:34:00.470.336.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.405562 in 1 packets - #/No: 18/18
      24.636908 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4358173 to 41.4356969
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999737 in 1 packets - #/No: 18/18
      24.9999059 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BE -> 32.4°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      BF -> 34.0°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180422_033328_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8628158 to 9.8631048
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.4055350  -- OK for SL --
   MIU time from 2018-04-22T03:33:28.426.999.000 to 2018-04-22T03:33:28.427.288.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.405562

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.2686398 to 17.2689288
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.6368150  -- OK for SL --
   MIU time from 2018-04-22T03:33:35.832.850.000 to 2018-04-22T03:33:35.833.139.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.636908

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.9057438 to 41.9060328
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-22T03:34:00.470.047.000 to 2018-04-22T03:34:00.470.336.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-22T03:33:28.426.858.750
   Max SL corrected packet time: 2018-04-22T03:34:00.470.195.750
   Min SL corrected XO rel time: 8.4270001
   Max SL corrected XO rel time: 40.4703377
   TM values from -29 to 24 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-22T03:33:28.426.858.750 (XO: 8.4270001, t=32.4°C)  Rank: 00002
   TM values: Max =     6  Min =   -15  Ave =    -4.47  Std =     2.92
   Min TM value:-15 at an offset of 103.475 usec -> 2018-04-22T03:33:28.426.962.225 (XO: 8.4271036)
   Max TM value:6 at an offset of 111.563 usec -> 2018-04-22T03:33:28.426.970.312 (XO: 8.4271116)
   Max dTM/dt [1/us]::1.40e+02 at an offset of 253.976 usec -> 2018-04-22T03:33:28.427.112.725 (XO: 8.4272541)
   Min dTM/dt [1/us]::-1.26e+02 at an offset of 214.101 usec -> 2018-04-22T03:33:28.427.072.850 (XO: 8.4272142)

 - Block number: 2 starting at 2018-04-22T03:33:35.832.709.750 (XO: 15.8328519, t=32.4°C)  Rank: 00002
   TM values: Max =     6  Min =   -14  Ave =    -4.48  Std =     2.69
   Min TM value:-14 at an offset of 27.688 usec -> 2018-04-22T03:33:35.832.737.437 (XO: 15.8328796)
   Max TM value:6 at an offset of 28.825 usec -> 2018-04-22T03:33:35.832.738.575 (XO: 15.8328807)
   Max dTM/dt [1/us]::1.18e+02 at an offset of 45.050 usec -> 2018-04-22T03:33:35.832.754.800 (XO: 15.8328969)
   Min dTM/dt [1/us]::-1.31e+02 at an offset of 304.826 usec -> 2018-04-22T03:33:35.833.014.576 (XO: 15.8331567)

 - Block number: 3 starting at 2018-04-22T03:34:00.469.906.750 (XO: 40.4700487, t=32.4°C)  Rank: 00009
   TM values: Max =    24  Min =   -29  Ave =    -4.49  Std =     5.45
   Min TM value:-29 at an offset of 107.413 usec -> 2018-04-22T03:34:00.470.014.162 (XO: 40.4701561)
   Max TM value:24 at an offset of 110.775 usec -> 2018-04-22T03:34:00.470.017.525 (XO: 40.4701595)
   Max dTM/dt [1/us]::2.86e+02 at an offset of 111.350 usec -> 2018-04-22T03:34:00.470.018.100 (XO: 40.4701600)
   Min dTM/dt [1/us]::-3.14e+02 at an offset of 134.863 usec -> 2018-04-22T03:34:00.470.041.613 (XO: 40.4701835)
