-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180421_092352_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.000088 to 30.022282
-each separately from 0s to 30s and from 88us to 22282us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.010616 in 1 packets - #/No: 18/18
      15.010711 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-21T09:23:44.000.088.000 to 2018-04-21T09:24:14.022.282.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.010616 in 1 packets - #/No: 18/18
      15.010711 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4355533 to 31.4354406
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999436 in 1 packets - #/No: 18/18
      14.9999437 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180421_092352_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4356429 to 1.4359319
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2018-04-21T09:23:44.000.088.000 to 2018-04-21T09:23:44.000.377.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010616

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4464909 to 16.4467799
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2018-04-21T09:23:59.010.993.000 to 2018-04-21T09:23:59.011.282.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010711

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4574349 to 31.4577239
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-21T09:24:14.021.993.000 to 2018-04-21T09:24:14.022.282.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-21T09:23:43.999.947.750
   Max SL corrected packet time: 2018-04-21T09:24:14.022.141.750
   Min SL corrected XO rel time: 0.0000896
   Max SL corrected XO rel time: 30.0222834
   TM values from -11 to 2 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-21T09:23:43.999.947.750 (XO: 0.0000896, t=34.0°C)  Rank: 00001
   TM values: Max =     1  Min =   -10  Ave =    -4.43  Std =     1.41
   Min TM value:-10 at an offset of 284.076 usec -> 2018-04-21T09:23:44.000.231.826 (XO: 0.0003737)
   Max TM value:1 at an offset of 19.088 usec -> 2018-04-21T09:23:43.999.966.837 (XO: 0.0001087)
   Max dTM/dt [1/us]::4.90e+01 at an offset of 5.213 usec -> 2018-04-21T09:23:43.999.952.962 (XO: 0.0000948)
   Min dTM/dt [1/us]::-4.40e+01 at an offset of 303.751 usec -> 2018-04-21T09:23:44.000.251.501 (XO: 0.0003934)

 - Block number: 2 starting at 2018-04-21T09:23:59.010.852.750 (XO: 15.0109940, t=34.0°C)  Rank: 00001
   TM values: Max =     1  Min =    -9  Ave =    -4.32  Std =     1.49
   Min TM value:-9 at an offset of 24.013 usec -> 2018-04-21T09:23:59.010.876.762 (XO: 15.0110181)
   Max TM value:1 at an offset of 70.263 usec -> 2018-04-21T09:23:59.010.923.012 (XO: 15.0110643)
   Max dTM/dt [1/us]::5.00e+01 at an offset of 82.188 usec -> 2018-04-21T09:23:59.010.934.937 (XO: 15.0110762)
   Min dTM/dt [1/us]::-5.00e+01 at an offset of 70.300 usec -> 2018-04-21T09:23:59.010.923.050 (XO: 15.0110643)

 - Block number: 3 starting at 2018-04-21T09:24:14.021.852.750 (XO: 30.0219944, t=34.0°C)  Rank: 00001
   TM values: Max =     2  Min =   -11  Ave =    -4.42  Std =     2.06
   Min TM value:-11 at an offset of 43.300 usec -> 2018-04-21T09:24:14.021.896.050 (XO: 30.0220377)
   Max TM value:2 at an offset of 4.450 usec -> 2018-04-21T09:24:14.021.857.200 (XO: 30.0219988)
   Max dTM/dt [1/us]::5.10e+01 at an offset of 43.675 usec -> 2018-04-21T09:24:14.021.896.425 (XO: 30.0220381)
   Min dTM/dt [1/us]::-5.20e+01 at an offset of 121.938 usec -> 2018-04-21T09:24:14.021.974.687 (XO: 30.0221163)
