-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180419_173248_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.047516 to 31.032785
-each separately from 0s to 31s and from 32496us to 266705us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.218611 in 1 packets - #/No: 18/18
      12.765791 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-19T17:32:40.047.516.000 to 2018-04-19T17:33:11.032.785.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.218611 in 1 packets - #/No: 18/18
      12.765791 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4370770 to 32.4369610
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999326 in 1 packets - #/No: 18/18
      12.9999514 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180419_173248_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4845944 to 1.4848834
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2185430  -- OK for SL --
   MIU time from 2018-04-19T17:32:40.047.516.000 to 2018-04-19T17:32:40.047.805.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.218611

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.7034264 to 19.7037154
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.7657430  -- OK for SL --
   MIU time from 2018-04-19T17:32:58.266.416.000 to 2018-04-19T17:32:58.266.705.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.765791

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.4694584 to 32.4697474
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-19T17:33:11.032.496.000 to 2018-04-19T17:33:11.032.785.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-19T17:32:40.047.375.750
   Max SL corrected packet time: 2018-04-19T17:33:11.032.644.750
   Min SL corrected XO rel time: 0.0475176
   Max SL corrected XO rel time: 31.0327865
   TM values from -28 to 19 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-19T17:32:40.047.375.750 (XO: 0.0475176, t=34.0°C)  Rank: 00003
   TM values: Max =    19  Min =   -28  Ave =    -4.53  Std =     6.06
   Min TM value:-28 at an offset of 59.563 usec -> 2018-04-19T17:32:40.047.435.312 (XO: 0.0475771)
   Max TM value:19 at an offset of 17.763 usec -> 2018-04-19T17:32:40.047.393.512 (XO: 0.0475353)
   Max dTM/dt [1/us]::2.50e+02 at an offset of 27.775 usec -> 2018-04-19T17:32:40.047.403.525 (XO: 0.0475454)
   Min dTM/dt [1/us]::-2.66e+02 at an offset of 59.525 usec -> 2018-04-19T17:32:40.047.435.275 (XO: 0.0475771)

 - Block number: 2 starting at 2018-04-19T17:32:58.266.275.750 (XO: 18.2664178, t=34.0°C)  Rank: 00003
   TM values: Max =    14  Min =   -23  Ave =    -4.61  Std =     4.99
   Min TM value:-23 at an offset of 95.475 usec -> 2018-04-19T17:32:58.266.371.225 (XO: 18.2665133)
   Max TM value:14 at an offset of 297.289 usec -> 2018-04-19T17:32:58.266.573.038 (XO: 18.2667151)
   Max dTM/dt [1/us]::2.20e+02 at an offset of 289.414 usec -> 2018-04-19T17:32:58.266.565.163 (XO: 18.2667072)
   Min dTM/dt [1/us]::-2.47e+02 at an offset of 297.201 usec -> 2018-04-19T17:32:58.266.572.951 (XO: 18.2667150)

 - Block number: 3 starting at 2018-04-19T17:33:11.032.355.750 (XO: 31.0324975, t=34.0°C)  Rank: 00003
   TM values: Max =    17  Min =   -27  Ave =    -4.61  Std =     5.49
   Min TM value:-27 at an offset of 111.588 usec -> 2018-04-19T17:33:11.032.467.337 (XO: 31.0326091)
   Max TM value:17 at an offset of 182.563 usec -> 2018-04-19T17:33:11.032.538.313 (XO: 31.0326801)
   Max dTM/dt [1/us]::2.51e+02 at an offset of 111.500 usec -> 2018-04-19T17:33:11.032.467.250 (XO: 31.0326090)
   Min dTM/dt [1/us]::-2.75e+02 at an offset of 111.563 usec -> 2018-04-19T17:33:11.032.467.312 (XO: 31.0326091)
