-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180417_163705_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.706163 to 40.038770
-each separately from 5s to 40s and from 38481us to 706452us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.387749 in 1 packets - #/No: 18/18
      10.943991 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-17T16:37:02.706.163.000 to 2018-04-17T16:37:37.038.770.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.387749 in 1 packets - #/No: 18/18
      10.943991 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9360439 to 41.9359151
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999116 in 1 packets - #/No: 18/18
      10.9999596 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180417_163705_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.6422055 to 7.6424945
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3876630  -- OK for SL --
   MIU time from 2018-04-17T16:37:02.706.163.000 to 2018-04-17T16:37:02.706.452.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.387749

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.0301575 to 31.0304465
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9439510  -- OK for SL --
   MIU time from 2018-04-17T16:37:26.094.201.000 to 2018-04-17T16:37:26.094.490.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.943991

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.9743975 to 41.9746865
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-17T16:37:37.038.481.000 to 2018-04-17T16:37:37.038.770.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-17T16:37:02.706.022.750
   Max SL corrected packet time: 2018-04-17T16:37:37.038.629.750
   Min SL corrected XO rel time: 5.7061642
   Max SL corrected XO rel time: 40.0387715
   TM values from -23 to 15 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-17T16:37:02.706.022.750 (XO: 5.7061642, t=32.4°C)  Rank: 00003
   TM values: Max =    11  Min =   -21  Ave =    -4.51  Std =     4.71
   Min TM value:-21 at an offset of 105.000 usec -> 2018-04-17T16:37:02.706.127.750 (XO: 5.7062692)
   Max TM value:11 at an offset of 129.838 usec -> 2018-04-17T16:37:02.706.152.587 (XO: 5.7062941)
   Max dTM/dt [1/us]::1.48e+02 at an offset of 129.800 usec -> 2018-04-17T16:37:02.706.152.550 (XO: 5.7062940)
   Min dTM/dt [1/us]::-1.75e+02 at an offset of 128.925 usec -> 2018-04-17T16:37:02.706.151.675 (XO: 5.7062931)

 - Block number: 2 starting at 2018-04-17T16:37:26.094.060.750 (XO: 29.0942023, t=32.4°C)  Rank: 00003
   TM values: Max =    14  Min =   -23  Ave =    -4.46  Std =     4.92
   Min TM value:-23 at an offset of 221.926 usec -> 2018-04-17T16:37:26.094.282.675 (XO: 29.0944243)
   Max TM value:14 at an offset of 99.513 usec -> 2018-04-17T16:37:26.094.160.262 (XO: 29.0943019)
   Max dTM/dt [1/us]::1.73e+02 at an offset of 99.200 usec -> 2018-04-17T16:37:26.094.159.950 (XO: 29.0943015)
   Min dTM/dt [1/us]::-1.94e+02 at an offset of 147.376 usec -> 2018-04-17T16:37:26.094.208.125 (XO: 29.0943497)

 - Block number: 3 starting at 2018-04-17T16:37:37.038.340.750 (XO: 40.0384825, t=32.4°C)  Rank: 00004
   TM values: Max =    15  Min =   -22  Ave =    -4.40  Std =     5.64
   Min TM value:-22 at an offset of 92.363 usec -> 2018-04-17T16:37:37.038.433.112 (XO: 40.0385749)
   Max TM value:15 at an offset of 184.601 usec -> 2018-04-17T16:37:37.038.525.350 (XO: 40.0386671)
   Max dTM/dt [1/us]::1.86e+02 at an offset of 171.013 usec -> 2018-04-17T16:37:37.038.511.763 (XO: 40.0386536)
   Min dTM/dt [1/us]::-1.78e+02 at an offset of 262.526 usec -> 2018-04-17T16:37:37.038.603.275 (XO: 40.0387451)
