-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180416_004627_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.943091 to 40.112016
-each separately from 1s to 40s and from 10995us to 943380us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.067615 in 1 packets - #/No: 18/18
      25.100443 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-16T00:46:19.943.091.000 to 2018-04-16T00:46:58.112.016.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.067615 in 1 packets - #/No: 18/18
      25.100443 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4337192 to 41.4335762
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999486 in 1 packets - #/No: 18/18
      24.9999084 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180416_004627_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3768081 to 3.3770971
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0675670  -- OK for SL --
   MIU time from 2018-04-16T00:46:19.943.091.000 to 2018-04-16T00:46:19.943.380.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.067615

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4446641 to 16.4449531
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.1003510  -- OK for SL --
   MIU time from 2018-04-16T00:46:33.010.995.000 to 2018-04-16T00:46:33.011.284.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.100443

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.5453041 to 41.5455931
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-16T00:46:58.111.727.000 to 2018-04-16T00:46:58.112.016.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-16T00:46:19.942.950.750
   Max SL corrected packet time: 2018-04-16T00:46:58.111.875.750
   Min SL corrected XO rel time: 1.9430924
   Max SL corrected XO rel time: 40.1120173
   TM values from -17 to 7 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-16T00:46:19.942.950.750 (XO: 1.9430924, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -16  Ave =    -4.47  Std =     3.13
   Min TM value:-16 at an offset of 284.676 usec -> 2018-04-16T00:46:19.943.235.426 (XO: 1.9433771)
   Max TM value:7 at an offset of 266.601 usec -> 2018-04-16T00:46:19.943.217.350 (XO: 1.9433590)
   Max dTM/dt [1/us]::1.27e+02 at an offset of 279.801 usec -> 2018-04-16T00:46:19.943.230.551 (XO: 1.9433722)
   Min dTM/dt [1/us]::-1.23e+02 at an offset of 142.738 usec -> 2018-04-16T00:46:19.943.093.488 (XO: 1.9432351)

 - Block number: 2 starting at 2018-04-16T00:46:33.010.854.750 (XO: 15.0109963, t=34.0°C)  Rank: 00001
   TM values: Max =     3  Min =   -12  Ave =    -4.45  Std =     2.13
   Min TM value:-12 at an offset of 37.238 usec -> 2018-04-16T00:46:33.010.891.987 (XO: 15.0110336)
   Max TM value:3 at an offset of 61.075 usec -> 2018-04-16T00:46:33.010.915.825 (XO: 15.0110574)
   Max dTM/dt [1/us]::8.30e+01 at an offset of 211.826 usec -> 2018-04-16T00:46:33.011.066.575 (XO: 15.0112082)
   Min dTM/dt [1/us]::-9.20e+01 at an offset of 0.875 usec -> 2018-04-16T00:46:33.010.855.625 (XO: 15.0109972)

 - Block number: 3 starting at 2018-04-16T00:46:58.111.586.750 (XO: 40.1117283, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -17  Ave =    -4.54  Std =     3.66
   Min TM value:-17 at an offset of 92.363 usec -> 2018-04-16T00:46:58.111.679.112 (XO: 40.1118207)
   Max TM value:7 at an offset of 86.975 usec -> 2018-04-16T00:46:58.111.673.725 (XO: 40.1118153)
   Max dTM/dt [1/us]::1.55e+02 at an offset of 156.201 usec -> 2018-04-16T00:46:58.111.742.950 (XO: 40.1118845)
   Min dTM/dt [1/us]::-1.62e+02 at an offset of 150.413 usec -> 2018-04-16T00:46:58.111.737.163 (XO: 40.1118787)
