-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180412_151743_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.355939 to 36.036998
-each separately from 10s to 36s and from 36709us to 360833us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.004316 in 1 packets - #/No: 18/18
      9.675876 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-12T15:17:43.454.939.000 to 2018-04-12T15:18:09.135.998.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.004316 in 1 packets - #/No: 18/18
      9.675876 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4381346 to 37.4380357
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999391 in 1 packets - #/No: 18/18
      9.9999620 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180412_151743_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.7940737 to 11.7943627
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.0042550  -- OK for SL --
   MIU time from 2018-04-12T15:17:43.454.939.000 to 2018-04-12T15:17:43.455.228.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.004316

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.7986177 to 27.7989067
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.6758390  -- OK for SL --
   MIU time from 2018-04-12T15:17:59.459.544.000 to 2018-04-12T15:17:59.459.833.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.675876

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.4747457 to 37.4750347
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-12T15:18:09.135.709.000 to 2018-04-12T15:18:09.135.998.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-12T15:17:43.454.798.750
   Max SL corrected packet time: 2018-04-12T15:18:09.135.857.750
   Min SL corrected XO rel time: 10.3559405
   Max SL corrected XO rel time: 36.0369991
   TM values from -25 to 25 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-12T15:17:43.454.798.750 (XO: 10.3559405, t=34.0°C)  Rank: 00007
   TM values: Max =    16  Min =   -25  Ave =    -4.50  Std =     4.44
   Min TM value:-25 at an offset of 186.663 usec -> 2018-04-12T15:17:43.454.985.413 (XO: 10.3561271)
   Max TM value:16 at an offset of 88.425 usec -> 2018-04-12T15:17:43.454.887.175 (XO: 10.3560289)
   Max dTM/dt [1/us]::1.71e+02 at an offset of 267.589 usec -> 2018-04-12T15:17:43.455.066.338 (XO: 10.3562080)
   Min dTM/dt [1/us]::-1.68e+02 at an offset of 183.263 usec -> 2018-04-12T15:17:43.454.982.013 (XO: 10.3561237)

 - Block number: 2 starting at 2018-04-12T15:17:59.459.403.750 (XO: 26.3605454, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -21  Ave =    -4.41  Std =     3.33
   Min TM value:-21 at an offset of 79.363 usec -> 2018-04-12T15:17:59.459.483.112 (XO: 26.3606247)
   Max TM value:13 at an offset of 124.350 usec -> 2018-04-12T15:17:59.459.528.100 (XO: 26.3606697)
   Max dTM/dt [1/us]::1.54e+02 at an offset of 124.163 usec -> 2018-04-12T15:17:59.459.527.912 (XO: 26.3606695)
   Min dTM/dt [1/us]::-1.24e+02 at an offset of 123.950 usec -> 2018-04-12T15:17:59.459.527.700 (XO: 26.3606693)

 - Block number: 3 starting at 2018-04-12T15:18:09.135.568.750 (XO: 36.0367101, t=34.0°C)  Rank: 00002
   TM values: Max =    25  Min =   -24  Ave =    -4.51  Std =     3.35
   Min TM value:-24 at an offset of 11.838 usec -> 2018-04-12T15:18:09.135.580.587 (XO: 36.0367220)
   Max TM value:25 at an offset of 65.313 usec -> 2018-04-12T15:18:09.135.634.062 (XO: 36.0367755)
   Max dTM/dt [1/us]::2.57e+02 at an offset of 65.263 usec -> 2018-04-12T15:18:09.135.634.012 (XO: 36.0367754)
   Min dTM/dt [1/us]::-2.96e+02 at an offset of 65.363 usec -> 2018-04-12T15:18:09.135.634.112 (XO: 36.0367755)
